First results obtained in France with the latest model of the Fresenius cell separator: AS 104.
In Besançon, we carried out 40 plateletphereses with the latest model of the Fresenius cell separator AS 104 to check this new system against the new generation of cell separators, according to the following criteria:  less than 2x10 6 leukocytes (before filtration) and more than 5x10 11 platelets.  The results show that platelet concentrates contained 5.04+/-0.88x10 11 platelets in a total volume of 435+/-113 mL.  The mean platelet recovery was 40.95+/-4.86% (from 31.7 to 51.6).  The leukocyte content was 2.28+/-5.48x10 6 and the red blood cell contamination was 3.48+/-2.38x10 8.  The quality of the platelets was very satisfactory.  There was no problem with donor biocompatibility or procedure safety, few adverse donor reactions (0.6%) and good therapeutic efficiency of platelet concentrates.
Coffe
C
C
Regional Blood Transfusion Centre, Besancon Cedex, France.
Couteret
Y
Y
Devillers
M
M
Fest
T
T
Hervé
P
P
Kieffer
Y
Y
Lamy
B
B
Masse
M
M
Morel
P
P
Pouthier-Stein
F
F
D002469:Cell Separation / Q000295:instrumentation* / Q000379:methods; D004869:Equipment Safety; D005260:Female; D006801:Humans; D007958:Leukocyte Count; D008297:Male; D008875:Middle Aged; D010976:Platelet Count; D010983:Plateletpheresis / Q000295:instrumentation* / Q000379:methods