Revision total hip arthroplasty using a cementless acetabular component: technique and results.
One hundred thirty-eight acetabular revisions for aseptic loosening were performed in 132 patients using a cementless hemispherical component coated with titanium mesh and inserted with supplemental screw fixation. Twelve patients died, 9 were lost to follow-up and 6 could not return, leaving 111 patients (115 hips) at mean 100 months follow-up. Thirteen hips required revision (11%); 4 for recurrent dislocations, 6 for sepsis, and 3 stable cups were revised (at the time of stem revision). No cup was revised for aseptic loosening. Radiographic review was available for 105 patients (109 hips) at mean 98 month follow-up (78 to 135 months). One hundred cups (91%) were considered stable. Seven cups were possibly unstable (7%), one cup was probably unstable (1%), and one septic hip had an unstable cup (1%). Four percent of the cups showed a complete radiolucency. A partial progressive radiolucency was seen in 3%, a partial nonprogressive radiolucency in 54%, and no radiolucency was present in 39%. A screw radiolucency was seen in 2%, and osteolysis at the cup margin in 4%. Revision of the acetabulum with a cementless porous-coated hemispherical fiber-metal component seems to be superior to the results reported for acetabular revisions with cement at similar follow-up.
Silverton
C D
CD
Department of Orthopaedic Surgery, Rush-Presbyterian-St. Luke's Medical Center, Chicago, IL 60612, USA.
Rosenberg
A G
AG
Sheinkop
M B
MB
Kull
L R
LR
Galante
J O
JO
D000077:Acetabulum / Q000601:surgery*; D000328:Adult; D000368:Aged; D001843:Bone Cements; D005260:Female; D005500:Follow-Up Studies; D006622:Hip Prosthesis; D006801:Humans; D008297:Male; D008875:Middle Aged; D011183:Postoperative Complications / Q000503:physiopathology; D011446:Prospective Studies; D011474:Prosthesis Design; D011475:Prosthesis Failure; D016059:Range of Motion, Articular; D012086:Reoperation / Q000295:instrumentation / Q000379:methods; D016896:Treatment Outcome