title: Multicenter evaluation of a new rapid automated human immunodeficiency virus antigen detection assay.
abstract: Although human immunodeficiency virus (HIV) antigen assays are of limited value for monitoring antiretroviral therapy, they play an important role for confirmatory testing of fourth generation HIV screening enzyme immunoassay (EIA) reactive samples. In a multicenter study, a new automated rapid p24 antigen assay, Elecsys HIV Ag (Roche Diagnostics Boehringer Mannheim GmbH, Penzberg, Germany), was compared to FDA licensed tests (Abbott HIV-1 Ag monoclonal and Coulter HIV-1 p24 antigen assay). In the evaluation 27 seroconversion panels were included, sera from the acute phase of infection, single and follow-up samples from HIV antibody positive patients, dilution series of HIV antigen positive standards, sera and cell culture supernatants infected with different HIV-1 subtypes (A-H, and O) HIV-2 and recombinant HIV-1 (gag/env) isolates. To challenge the specificity of the new assay, 2565 unselected blood donors, sera from pregnant women, dialysis and hospitalized patients and 407 potentially cross-reactive samples were investigated. Acute HIV infection was detected in three to eight seroconversion panels earlier with Elecsys HIV Ag than with the alternative assays. Higher numbers of serum samples from HIV infected patients tested positive by Elecsys HIV Ag than with the comparative assays. All HIV-1 subtypes and HIV-2 isolates were recognized with Elecsys HIV Ag. Abbott HIV-1 Ag monoclonal and Coulter HIV-1 p24 antigen assay showed a variable sensitivity for the different HIV-1 subtypes. The specificity of Elecsys HIV Ag and Coulter HIV-1 p24 antigen assay were 99.8 and 99.93%, respectively. All the eight sera that were false reactive by Elecsys HIV Ag were tested negative with the Elecsys HIV Ag Neutralization Test. In conclusion, Elecsys HIV Ag was more sensitive than the alternative assays and showed a high specificity in combination with the neutralization assay. The very short incubation time of 18 min and the fully automated procedure of Elecsys HIV Ag which permits direct testing from the primary patient blood collection tube, represent a major improvement for routine laboratory diagnosis in comparison to the alternative assays.
authors: Laboratoires Réunis Kutter-Lieners-Hastert, Centre Langwies, Junglinster, Luxembourg. laborklh@pt.lu
authors: B
authors: B
authors: Weber
authors: A
authors: A
authors: Mühlbacher
authors: U
authors: U
authors: Michl
authors: G
authors: G
authors: Paggi
authors: V
authors: V
authors: Bossi
authors: C
authors: C
authors: Sargento
authors: R
authors: R
authors: Camacho
authors: E H
authors: EH
authors: Fall
authors: A
authors: A
authors: Berger
authors: U
authors: U
authors: Schmitt
authors: W
authors: W
authors: Melchior
mesh_terms: D000911:Antibodies, Monoclonal / Q000276:immunology; D005069:Evaluation Studies as Topic; D005260:Female; D016655:HIV Core Protein p24 / Q000097:blood*; D015658:HIV Infections / Q000175:diagnosis* / Q000821:virology; D015497:HIV-1 / Q000302:isolation & purification*; D006801:Humans; D007118:Immunoassay / Q000379:methods*; D009500:Neutralization Tests; D011247:Pregnancy; D012367:RNA, Viral / Q000032:analysis; D011933:Reagent Kits, Diagnostic; D020133:Reverse Transcriptase Polymerase Chain Reaction; D012680:Sensitivity and Specificity
chemical_list: D000911:Antibodies, Monoclonal; D016655:HIV Core Protein p24; D012367:RNA, Viral; D011933:Reagent Kits, Diagnostic