Reliability of a new device to assess the oxygen consumption of human respiratory muscles.
OBJECTIVE
This study tests the reliability of a new device for assessing the oxygen consumption of the respiratory muscles (VO2 resp.).


METHODS
Fourteen healthy male volunteers participated in the study. The device consists of an expandable external ventilatory dead space created with pieces of plastic tubing and a spirometer filled with 100% oxygen. It also incorporates a carbon dioxide absorber. Total VO2 (VO2 tot.) was recorded from the spirometric closed circuit and ventilation (V(E)), from the spirometer tracing. For each subject the test procedure was carried out in duplicate (T1 and T2) after an overnight fast. The dead space was increased at a constant rate of 260 mL every 90 s, and VO2 tot. and V(E) increased progressively. Because log VO2 tot. was linearly related to V(E), we calculated the slope value (log VO2-V(E)) and the Y-intercept (VE = 0) of the semilog regression representing, respectively, VO2 resp. and metabolic VO2 (VO2 met.).


RESULTS
When compared with values in the literature, these values did not differ from those recorded in subjects of a similar age group. The VO2 resp. and VO2 met. calculated in T1 and T2 were not different (VO2 resp. = 0.0066 +/- 0.0005 for T1 vs 0.0067 +/- 0.0005 log mL x min(-1)/L x min(-1) for T2 and VO2 met. = 269.3 +/- 28.6 for T1 vs 281.9 +/- 24.1 mL x min(-1) for T2). The coefficients of variation were: 25% at T1 and 23% at T2 for VO2 resp. and 34% at T1 and 29% at T2 for VO2 met. Moreover, significant correlations (r = 0.96, P < 0.001 for VO2 resp., r = 0.95, P < 0.001 for VO2 met.), high coefficients of determination (r2 = 0.92 for VO2 resp., r2 = 0.90 for VO2 met.) and negligible SEE (0.0005 for VO2 resp., 0.2 mL x min(-1) for VO2 met.) were found between the two tests. When we plotted the mean values of VO2 resp. and VO2 met. measured at T1 and T2 against their respective differences, more than 95% of the slight differences ranged between the limits defined by mean value +/- 2 SD, reflecting the small discrepancy between duplicate measurements.


CONCLUSIONS
The results confirm that the test performed with this device is useful and reliable for assessing the VO2 resp. in healthy subjects.
TEF-Adaptations Physiologiques et Comportementales, Faculté de Médecine (EA 2088), Université de Picardie Jules Verne, Amiens, France. Said.Ahmaidi@ca.u-picardie.fr
S
S
Ahmaidi
D
D
Comte
N
N
Topin
M
M
Hayot
S
S
Delanaud
M
M
Ramonatxo
N
N
His
G
G
Vardon
M
M
Freville
J P
JP
Libert
C
C
Préfaut
D000328:Adult; D004867:Equipment Design; D006801:Humans; D016014:Linear Models; D008297:Male; D008991:Monitoring, Physiologic / Q000295:instrumentation*; D010101:Oxygen Consumption / Q000502:physiology*; D015203:Reproducibility of Results; D012129:Respiratory Function Tests; D012132:Respiratory Muscles / Q000378:metabolism*