Continuous monitoring of jugular bulb oxyhaemoglobin saturation using the Edslab dual lumen oximetry catheter during and after cardiac surgery.
Fibreoptic jugular bulb oximetry has been validated for use in the care of severely head-injured patients. We compared bench and fibreoptic methods of measuring jugular bulb oxyhaemoglobin saturation (SjO2) in 33 patients undergoing cardiac surgery both during cardiopulmonary bypass (CPB) and in the early postoperative period. After insertion of a fibreoptic reflectance oximetry catheter into the jugular bulb, it was calibrated against a bench oximeter. Comparisons were made while on CPB (n = 60) and in the postoperative period for up to 18 h (n = 215). There was negligible bias throughout. There were wide limits of agreements (mean difference +/- 2SD) between the two methods during operation (-20.29% to 18.05%), whereas after operation the limits of agreement were far narrower (-6.39% and 7.45%). Measurement of SjO2 by the fibreoptic method compared poorly with bench oximetry during CPB but there was good agreement between the two methods in the early postoperative period.
Millar
S A
SA
Department of Anaesthetics, University of Edinburgh, Royal Infirmary of Edinburgh, UK.
Alston
R P
RP
Souter
M J
MJ
Andrews
P J
PJ
D000328:Adult; D002315:Cardiopulmonary Bypass; D001026:Coronary Artery Bypass; D005336:Fiber Optic Technology; D019918:Heart Valve Prosthesis Implantation; D006801:Humans; D007601:Jugular Veins; D016343:Monitoring, Intraoperative / Q000379:methods; D010092:Oximetry / Q000379:methods*; D010108:Oxyhemoglobins / Q000378:metabolism*; D011182:Postoperative Care / Q000379:methods
D010108:Oxyhemoglobins