Reoperations for acute prosthetic thrombosis and pannus: an assessment of rates, relationship and risk.
OBJECTIVE
Mechanical valvular prostheses have the advantage of longevity but carry a risk of thrombosis which is dependant on valve design, materials and host-related interface. While pannus is common to both biologic and mechanical valves, acute prosthetic thrombosis is mostly a complication of mechanical valves; therefore we investigated to find rates and risk of these obstructive complications.


METHODS
Between 1/1/70 and 31/12/97, 2680 patients received at least one mechanical prosthesis in the aortic or mitral or tricuspid position and a total of 3014 operations were performed. Follow-up included 18523 years and was 98% complete. Incidence rates, Kaplan-Meier estimates, modeling of the hazard and multivariate analysis in the hazard domain were used in the analysis.


RESULTS
Overall survival was 76%, 64%, 51%, 38.5% and 29% at 5, 10, 15, 20 and 25 years, respectively. It was significantly better in aortic than in mitral than in double prosthesis. 290 patients received a single reoperation, 37 a second, six a third and one a fourth reoperation. Two-hundred and fifty-one of these reoperations were exclusively due to malfunction of mechanical prosthesis, nine to malfunction of both mechanic and biologic prostheses. Most frequent reoperative indications was dehiscence (133), pannus (48) and thrombosis (29). The linearized rate of reoperations for pannus was 0.24%/patient per year, for valvular thrombosis 0.15%/patient per year. The shape of the thrombotic hazard was constant (at random) and the relative risk 12 times higher for tricuspid prosthesis, seven times higher for mitral prosthesis. Multivariate analysis controlling for prosthetic position, age, sex and prosthetic size, showed a 67% risk reduction with larger prosthesis (>27 mm), a 69% risk reduction with the Sorin tilting disk prosthesis and an 83% risk reduction with the bileaflet prosthesis. Pannus hazard shows a delayed exponential rise and was two times higher in tricuspid and three times higher in mitral position. Multivariate analysis showed a 50% risk reduction with larger prosthesis, an 11 times higher hazard of old (caged-disk, caged ball) prosthesis and a three times higher hazard of Lillehei-Kaster prosthesis. Reoperation for thrombosis has a 62% perioperative (30 days) survival compared to 92% survival of pannus reoperation.


CONCLUSIONS
Mechanical valves have a low incidence of reoperation, mostly for prosthetic dehiscence. Pannus development is the next frequent complication increasing with time since implant, therefore in this series it was related to old valvular models and tilting disk prosthesis, with longer follow-up. Acute thrombosis occurs significantly earlier than pannus formation. Despite shorter follow-up we are therefore confident that bileaflet prostheses are less prone to this complication and pannus is a rare early etiologic factor. Thrombosis has very high operative risk as compared to pannus, justifying the present trend to thrombolysate selected cases.
Rizzoli
G
G
Istituto Chirurgia Cardiovascolare, University of Padova, Padua, Italy. rzzgli@ux1.unipd.it
Guglielmi
C
C
Toscano
G
G
Pistorio
V
V
Vendramin
I
I
Bottio
T
T
Thiene
G
G
Casarotto
D
D
D000208:Acute Disease; D001021:Aortic Valve; D001705:Bioprosthesis; D006350:Heart Valve Prosthesis; D019918:Heart Valve Prosthesis Implantation / Q000009:adverse effects*; D006801:Humans; D008943:Mitral Valve; D011474:Prosthesis Design; D012086:Reoperation; D012189:Retrospective Studies; D018570:Risk Assessment; D013927:Thrombosis / Q000209:etiology*; D014261:Tricuspid Valve