Evaluation and field calibration of the Miniram PDM-3 aerosol monitor for measuring respirable and total coal dust.
The MIE Miniram PDM-3 is a real-time aerosol dust monitor designed to measure dust based on Mie scattering. It has an optional in-line filter that, when attached to a constant flow air sampling pump, allows a gravimetric air sample to be collected from the same air stream that passes through the Miniram sensing chamber. This study compared real-time Miniram digital respirable and total dust readings with concentrations from an in-line 5 microns polyvinyl chloride filter connected to a constant flow pump and with results from traditional respirable and total dust samples. Area samples were collected at three coal-fired power generating facilities over a 2-month period. Traditional respirable dust concentrations ranged from 0.04 to 10.8 mg/m3. In the monitored range of concentrations the Miniram respirable concentrations and the in-line respirable dust concentrations were not significantly different (p > or = 0.05), nor were the Miniram in-line filter concentrations and the side-by-side respirable dust concentrations significantly different (p > or = 0.05). However, the Miniram respirable concentrations and the traditional respirable concentrations were significantly different (p = 0.02). The Statistical Analysis System (SAS) procedure CALIS, which meets the statistical requirements for developing calibration lines for two variables each measured with error, is used to develop field calibration curves for the comparisons between direct-reading instrument concentrations and concentrations from traditional National Institute for Occupational Safety and Health methods for total and respirable dust.
Middendorf
P J
PJ
Georgia Tech Research Institute, Georgia Institute of Technology, Atlanta 30332, USA.
Lehocky
A H
AH
Williams
P L
PL
D000336:Aerosols / Q000032:analysis; D000393:Air Pollutants / Q000032:analysis*; D016902:Air Pollution, Indoor / Q000517:prevention & control; D002138:Calibration; D003031:Coal; D004391:Dust / Q000032:analysis*; D004784:Environmental Monitoring / Q000295:instrumentation*; D004867:Equipment Design; D006801:Humans; D016273:Occupational Exposure / Q000517:prevention & control
D000336:Aerosols; D000393:Air Pollutants; D003031:Coal; D004391:Dust