Standardized visual evoked potentials for telematic electrodiagnosis from five laboratories in three European countries.
The pattern-reversal visual evoked potential (PR VEP) is an electrodiagnostic test which uses an alternating checkerboard pattern as the visual stimulus. To validate the performance of an automated testing system, the equipment was installed in five locations in three countries (the UK, Greece and Cyprus). PR VEPs require accurate control of stimulus size, luminance, contrast, timing, multichannel recordings and various modes of signal processing. Recordings were obtained from 70 subjects, 18-34 years old (seven male and seven female in each laboratory). The data were transmitted by telephone line to the central station in Bristol. Only in Bristol were the operators experienced. There were no significant differences between laboratories in latency or amplitude of the N70 and P100 components. There were no significant differences between eyes. There were significant differences between the sexes but this was related to head size. Decreased square size used in stimulation produced an increased latency in both components. The amplitude of the N70 component increased while that of the P100 decreased with reduced square size. The results confirmed the ability of the automated system to produce standardized PR VEP recordings comparable to those obtainable in advanced laboratories. The system performed well in different locations using inexperienced operators. The system is suitable for multicentre research studies and telemedicine.
Papakostopoulos
D
D
Electrodiagnostic Department, Bristol Eye Hospital, UK. az24@cityscape.co.uk
Hart
J C
JC
Papakostopoulos
S
S
Dodson
K
K
Williams
A
A
D000293:Adolescent; D000328:Adult; D004568:Electrodiagnosis; D005074:Evoked Potentials, Visual; D005128:Eye Diseases / Q000175:diagnosis*; D005260:Female; D006801:Humans; D008297:Male; D017216:Telemedicine / Q000379:methods*; D013686:Telemetry / Q000295:instrumentation / Q000379:methods*