Laser desorption in transmission geometry inside a Fourier-transform ion cyclotron resonance mass spectrometer.
We report here the first application of laser desorption (LD) in transmission geometry (backside irradiation of the sample through a transparent support) inside a Fourier-transform ion cyclotron resonance mass spectrometer (FT-ICR). A probe-mounted fiber optic assembly was used to simplify the implementation of this LD technique. This setup requires little or no instrument modifications, has minimum maintenance requirements, and is relatively inexpensive to build. The performance of the probe was tested by determining the molecular weight of a commercial polystyrene standard from its matrix-assisted laser desorption/ionization (MALDI) spectrum. The measured average molecular weight is comparable to that obtained for the same sample by MALDI in the conventional top-illumination arrangement (reflection geometry) and by the manufacturer of the sample by gel permeation chromatography. The average velocities measured for ions evaporated by transmission mode LD of several neat samples are about half the velocity of those obtained by using the reflection geometry. Therefore, transmission mode irradiation of the sample holds promise to desorb ions that are easier to trap in an ICR cell. An oscillating capillary nebulizer was adapted for the deposition of analytes to improve sampling reproducibility.
Pérez
J
J
Department of Chemistry, Purdue University, West Lafayette, Indiana 47907-1393, USA. Hilkka@purdue.edu
Petzold
C J
CJ
Watkins
M A
MA
Vaughn
W E
WE
Kenttämaa
H I
HI
D002850:Chromatography, Gel; D003460:Crystallization; D017357:Cyclotrons; D005336:Fiber Optic Technology; D005583:Fourier Analysis; D008970:Molecular Weight; D055100:Optical Fibers; D011137:Polystyrenes / Q000737:chemistry; D019032:Spectrometry, Mass, Matrix-Assisted Laser Desorption-Ionization / Q000379:methods*
D011137:Polystyrenes