Variations in the feedback of hearing aids.
Variations in the loop response of hearing aids caused by jaw movements, variations in acoustics outside the ear, and variations of vent size have been identified. Behind The Ear (BTE) and In The Ear Canal (ITEC) hearing aids were considered. The largest variations among the variations of the acoustics outside the ear, except when the hearing aid was partly removed, were found with the ITEC when a telephone set was placed by the ear. The variations of the loop response caused by changes in vent size were compared with the variations of a theoretical model of the feedback path. The theoretical model was also used to compare the feedback of different designs of the vent that gives the same acoustic impedance at low frequencies. The calculated feedback was less with the short vents (12 mm) than the long vents (24 mm).
Hellgren
J
J
Department of Neuroscience and Locomotion, Linköping University, Sweden.
Lunner
T
T
Arlinger
S
S
D000162:Acoustics; D001307:Auditory Perception / Q000502:physiology*; D005246:Feedback; D006310:Hearing Aids; D006801:Humans; D007568:Jaw / Q000502:physiology*; D009068:Movement / Q000502:physiology*