title: Successful repair of a ventricular assist system percutaneous lead.
abstract: A patient with an implanted, electrically powered, ventricular assist device (Thermo Cardiosystems VE HeartMate) experienced a partial break of the percutaneous lead 5 months after implantation. The break (limited to the Silicone rubber tube) occurred at the junction of the lead with the Y-connector to the controller and vent, leaving approximately 5 cm of exposed lead from the skin exit site to the connector. Electronic and pumping functions of the pump continued, but the opening in the lead (which went more that half way around the circumference) prevented the use of pneumatic actuation as a back-up mode for pump operation, and placed the pump at risk for contamination. Repair of the lead without surgical intervention was desirable, with ease of repair and minimal risk to the patient being the top priorities. The use of multiple layers of heat-shrink tubing or external metal stents was ruled out in favor of a three stage repair procedure. The first stage involved the removal of the Dacron velour in-growth material from the lead to expose the underlying Silicone rubber tube. While the opening in the tube was held shut, a coating of medical grade Silicone rubber adhesive was applied to the tube, then wrapped with a woven Dacron mesh, followed by two layers of plastic wrapping material to protect the adhesive. This initial layer was secured by an external stent of tubing with cable ties. After several days to allow for complete curing of the adhesive, the adhesive coating with mesh was repeated. The final step involved a double layer wrap of a 1 mm thick Silicone rubber sheeting with mesh incorporation and adhesive secured in place with cable ties. After completion of the repair and verification of the ability to operate the device with pneumatic actuation, the patient was discharged with no recurrence of the problem after 8 months of weekly follow-up. This experience demonstrates the need to clinically anticipate component repair or replacement without total device replacement in future implantable blood pump systems.
authors.affiliation: Utah Artificial Heart Program, The Department of Cardiovascular Services, LDS Hospital, Salt Lake City 84103, USA.
authors.forename: G M
authors.initials: GM
authors.lastname: Pantalos
authors.forename: J D
authors.initials: JD
authors.lastname: Marks
authors.forename: E E
authors.initials: EE
authors.lastname: Richardson
authors.forename: K E
authors.initials: KE
authors.lastname: Nelson
authors.forename: J W
authors.initials: JW
authors.lastname: Long
mesh_terms: D000269:Adhesives; D000328:Adult; D001424:Bacterial Infections / Q000517:prevention & control; D009202:Cardiomyopathies / Q000628:therapy*; D004866:Equipment Contamination; D005260:Female; D006353:Heart-Assist Devices; D006801:Humans; D011475:Prosthesis Failure; D012828:Silicones; D015607:Stents
chemical_list: D000269:Adhesives; D012828:Silicones