title: Semi-automatic analysis of fire debris
abstract: Automated analysis of fire residues involves a strategy which deals with the wide variety of received criminalistic samples. Because of unknown concentration of accelerant in a sample and the wide range of flammable products, full attention from the analyst is required. Primary detection with a photoionisator resolves the first problem, determining the right method to use: the less responsive classical head-space determination or absorption on active charcoal tube, a better fitted method more adapted to low concentrations can thus be chosen. The latter method is suitable for automatic thermal desorption (ATD400), to avoid any risk of cross contamination. A PONA column (50 mx0.2 mm i.d.) allows the separation of volatile hydrocarbons from C(1) to C(15) and the update of a database. A specific second column is used for heavy hydrocarbons. Heavy products (C(13) to C(40)) were extracted from residues using a very small amount of pentane, concentrated to 1 ml at 50 degrees C and then placed on an automatic carousel. Comparison of flammables with referenced chromatograms provided expected identification, possibly using mass spectrometry. This analytical strategy belongs to the IRCGN quality program, resulting in analysis of 1500 samples per year by two technicians.
authors.affiliation: Institut de Recherche Criminelle de la Gendarmerie Nationale, Fort de Rosny, 1 Boulevard Theophile Sueur, 93111, Rosny-sous-Bois, France.
authors.initials: P
authors.lastname: Touron
authors.initials: P
authors.lastname: Malaquin
authors.initials: D
authors.lastname: Gardebas
authors.initials: J
authors.lastname: Nicolai