title: Micromachined ultrasound transducers with improved coupling factors from a CMOS compatible process
abstract: For medical high frequency acoustic imaging purposes the reduction in size of a single transducer element for one-dimensional and even more for two-dimensional arrays is more and more limited by fabrication and cabling technology. In the fields of industrial distance measurement and simple object recognition low cost phased arrays are lacking. Both problems can be solved with micromachined ultrasound transducers (MUTs). A single transducer is made of a large number of microscopic elements. Because of the array structure of these transducers, groups of elements can be built up and used as a phased array. By integrating parts of the sensor electronics on chip, the cabling effort for arrays can be reduced markedly. In contrast to standard ultrasonic technology, which is based on massive thickness resonators, vibrating membranes are the radiating elements of the MUTs. New micromachining technologies have emerged, allowing a highly reproducible fabrication of electrostatically driven membranes with gap heights below 500 nm. A microelectronic BiCMOS process was extended for surface micromechanics (T. Scheiter et al., Proceedings 11th European Conference on Solid-State Transducers, Warsaw, Vol. 3, 1997, pp. 1595-1598). Additional process steps were included for the realization of the membranes which form sealed cavities with the underlying substrate. Membrane and substrate are the opposite electrodes of a capacitive transducer. The transducers can be integrated monolithically on one chip together with the driving, preamplifying and multiplexing circuitry, thus reducing parasitic capacities and noise level significantly. Owing to their low mass the transducers are very well matched to fluid loads, resulting in a very high bandwidth of 50-100% (C. Eccardt et al., Proceedings Ultrasonics Symposium, San Antonio, Vol. 2, 1996, pp. 959-962; P.C. Eccardt et al., Proceedings of the 1997 Ultrasonics Symposium, Toronto, Vol. 2, 1997, pp. 1609-1618). In the following it is shown how the BiCMOS process has been modified to meet the demands for ultrasound generation and reception. Bias and driving voltages have been reduced down to the 10 V range. The electromechanical coupling is now almost comparable with that for piezoelectric transducers. The measurements exhibit sound pressures and bandwidths that are at least comparable with those of conventional piezoelectric transducer arrays.
authors.affiliation: Siemens AG, ZT KM 2, Munchen, Germany. peter-christian.eccardt@mchp.siemens.de
authors.initials: PC
authors.lastname: Eccardt
authors.initials: K
authors.lastname: Niederer