Good design practice for medical devices and equipment, Part II: Design for validation.
Medical devices and their associated process equipment must be reliable and fit for purpose. In light of the recent changes to the medical device regulations, manufacturers must now take an integrated approach to design, development and validation. Good design practice encourages this integrated approach while ensuring fitness for purpose within commercial reality. A review of current literature related to good design practice carried out in Part I of this paper showed that there is inadequate guidance regarding the integration of validation with design. This paper proposes a practical approach to design for validation aimed at making devices easier and more economic to validate. The approach comes in the form of a model of design for validation that illustrates the basic relationship between design, development and validation and a series of design tactics that were formulated in order to help designers take a more proactive approach to validation during design.
Alexander
K
K
Cambridge Engineering Design Centre, Department of Engineering, University of Cambridge, UK.
Clarkson
P J
PJ
D001698:Biomedical Engineering; D004867:Equipment Design / Q000592:standards; D006801:Humans; D008962:Models, Theoretical; D018570:Risk Assessment