Eccentricities of dual chamber implantable pacemakers.
The consequences of a short 160 ms postventricular atrial refractory period (PVARP) were studied in 27 patients with Medtronic Elite series of pacemakers (programmed to the unipolar mode) by using chest wall stimulation and myopotential interference. Because the ventricular refractory period is nonprogrammable at 230 ms, appropriately timed signals produced As-Vr combinations where As is an atrial sensed event and Vr is a ventricular event detected in the ventricular refractory period. The As-Vr combination inhibited the release of the atrial stimulus at the completion of the atrial escape interval in the DDI, DDIR, DDD, and DDDR modes. A paced ventricular beat (Vp) was documented to initiate automatic PVARP extension in circumstances where Vp was preceded by a Vr event and the pacemaker interpreted the Vr-Vp sequence as 2 ventricular events without an intervening atrial event and Vp as a ventricular premature beat. Knowledge of these eccentricities should facilitate the interpretation of complex pacemakers responses to interference.
Barold
S S
SS
Genesee Hospital, Rochester, New York, USA. ssbarold@aol.com
D002304:Cardiac Pacing, Artificial / Q000379:methods*; D006321:Heart / Q000502:physiology*; D006329:Heart Conduction System / Q000502:physiology; D006801:Humans; D010138:Pacemaker, Artificial