title: Capillary electrophoresis separations on a planar chip with the column-coupling configuration of the separation channels
abstract: Some basic aspects of capillary electrophoresis (CE) separations on a poly(methyl methacrylate) chip provided with two separation channels in the column-coupling (CC) configuration and on-column conductivity detectors were studied. The CE methods employed in this study included isotachophoresis (ITP), capillary zone electrophoresis (CZE), and CZE with on-line ITP sample pretreatment (ITP-CZE). Hydrodynamic and electroosmotic flows of the solution in the separation compartment of the chip were suppressed, and electrophoresis was a dominant transport process in the separations performed by these methods. Very reproducible migration velocities of the separated constituents were typical under such transport conditions, and consequently, test analytes could be quantified by various ITP techniques with 1-2% RSD. The CC configuration of the separation channels provides means for an effective combination of an enhanced load capacity of the separation system with high detection sensitivities for the analytes in concentration-cascade ITP separations. In this way, for example, succinate, acetate, and benzoate could be separated also in instances when they were present in the loaded sample (1.2 microL) at 1 mmol/L concentrations while their limits of detection ranged from 8 to 12 micromol/L concentrations. A well-defined ITP concentration of the analyte(s) combined with an in-column sample cleanup (via an electrophoretically driven removal of the matrix constituents from the separation compartment) can be integrated into the separations performed on the CC chip. These sample pretreatment capabilities were investigated in ITP-CZE separations of model samples in which nitrite, phosphate, and fluoride (each at a 10 micromol/L concentration) accompanied matrix constituents (sulfate and chloride) at considerably higher concentrations. Here, both the concentration of the analytes and cleanup of the sample were included in the ITP separation in the first separation channel while the second separation channel served for the CZE separation of the ITP pretreated sample and the detection of the analytes.
authors.affiliation: Department of Analytical Chemistry, Faculty of Natural Sciences, Comenius University, Bratislava, Slovak Republic. kaniansky@fns.uniba.sk
authors.initials: D
authors.lastname: Kaniansky
authors.initials: M
authors.lastname: Masar
authors.initials: J
authors.lastname: Bielcikova
authors.initials: F
authors.lastname: Ivanyi
authors.initials: F
authors.lastname: Eisenbeiss
authors.initials: B
authors.lastname: Stanislawski
authors.initials: B
authors.lastname: Grass
authors.initials: A
authors.lastname: Neyer
authors.initials: M
authors.lastname: Johnck