title: A new G-M counter dead time model
abstract: A hybrid G-M counter dead time model was derived by combining the idealized paralyzable and non-paralyzable models. The new model involves two parameters, which are the paralyzable and non-paralyzable dead times. The dead times used in the model are very closely related to the physical dead time of the G-M tube and its resolving time. To check the validity of the model, the decaying source method with 56Mn was used. The corrected counting rates by the new G-M dead time model were compared with the observed counting rates obtained from the measurement and gave very good agreement within 5% up to 7 x 10(4) counts/s for a G-M tube with a dead time of about 300 micros.
authors.affiliation: Center for Engineering Applications of Radioisotopes, North Carolina State University, Raleigh 27695-7909, USA. slee2@eos.ncsu.edu
authors.initials: SH
authors.lastname: Lee
authors.initials: RP
authors.lastname: Gardner