Standards for oesophageal manometry. A position statement from the Gruppo Italiano di Studio Motilità Apparato Digerente (GISMAD).
Manometry is an important tool in the diagnosis of oesophageal motility, disorders, but proper instruments and methods are needed to obtain useful clinical information. The authors reviewed the minimal technical requirements, operative aspects, which information the final report should contain as well as indications and contraindications of the text itself. Technical requirements: At least a three-channel, multiple-lumen catheter perfused with a pneumo-hydraulic capillary infusion system which ensures deltaP/deltaT>150-200 mmHg/sec.; data should be recorded at a sampling rate of > or =8 Hz to study the oesophageal body and lower oesophageal sphincter; lower oesophageal sphincter tonic (pressure) and phasic activity (relaxations) and oesophageal body amplitude and peristaltic activity should be recorded. The final report must contain the patient's details, the indication for the test and a manometric diagnosis. Indications for manometry: Dysphagia (after ruling out any organic pathology); non- cardiac chest pain (after ruling out any cardiopulmonary involvement); systemic collagenosis (to investigate oesophageal involvement); gastro-oesophageal reflux disease (if surgery is planned). Contraindications are limited to: pharyngeal or upper oesophageal obstructions, oesophageal bullous disorder, cardiac conditions in which vagal stimulation may not be tolerated, severe coagulopathy and patient non-compliance.
Passaretti
S
S
Gastroenterology Service, Ospedale S. Raffaele, Milan, Italy.
Zaninotto
G
G
Di Martino
N
N
Leo
P
P
Costantini
M
M
Baldi
F
F
D004867:Equipment Design; D015154:Esophageal Motility Disorders / Q000175:diagnosis* / Q000503:physiopathology; D004947:Esophagus / Q000503:physiopathology*; D006801:Humans; D007558:Italy; D008365:Manometry / Q000592:standards*; D011312:Pressure