title: Protein mapping by two-dimensional high performance liquid chromatography.
abstract: Current developments in drug discovery in the pharmaceutical industry require highly efficient analytical systems for protein mapping providing high resolution, robustness, sensitivity, reproducibility and a high throughput of samples. The potential of two-dimensional (2D) HPLC as a complementary method to 2D-gel electrophoresis is investigated, especially in view of speed and repeatability. The method will be applied for proteins of a molecular mass <20 000 which are not well resolved in 2D-gel electrophoresis. The 2D-HPLC system described in this work consisted of anion- or cation-exchange chromatography in the first dimension and reversed-phase chromatography in the second dimension. We used a comprehensive two-dimensional approach based on different separation speeds. In the first dimension 2.5 microm polymeric beads bonded with diethylaminoethyl and sulfonic acid groups, respectively, were applied as ion exchangers and operated at a flow-rate of 1 ml/min. To achieve very high-speed and high-resolution separations in the second dimension, short columns of 14 x 4.6 mm I.D. with 1.5 microm n-octadecyl bonded, non-porous silica packings were chosen and operated at a flow-rate of 2.5 ml/min. Two reversed-phase columns were used in parallel in the second dimension. The analyte fractions from the ion-exchange column were transferred alternatively to one of the two reversed-phase columns using a 10-port switching valve. The analytes were deposited in an on-column focusing mode on top of one column while the analytes on the second column were eluted. Proteins, which were not completely resolved in the first dimension can, in most cases, be baseline-separated in the second dimension. The total value of peak capacity was calculated to 600. Fully unattended overnight runs for repeatability studies proved the applicability of the system. The values for the relative standard deviation (RSD) of the retention times of proteins were less than 1% (n = 15), while the RSDs of the peak areas were less than 15% (n = 15) on average. The limit of detection was 300 ng of protein on average and decreased to 50 ng for ovalbumin. The 2D-HPLC system offered high-resolution protein separations with a total analysis time of less than 20 min, equivalent to the run time of the first dimension.
authors.affiliation: Institut für Anorganische Chemie und Analytische Chemie, Johannes Gutenberg-Universität, Mainz, Germany.
authors.forename: K
authors.initials: K
authors.lastname: Wagner
authors.forename: K
authors.initials: K
authors.lastname: Racaityte
authors.forename: K K
authors.initials: KK
authors.lastname: Unger
authors.forename: T
authors.initials: T
authors.lastname: Miliotis
authors.forename: L E
authors.initials: LE
authors.lastname: Edholm
authors.forename: R
authors.initials: R
authors.lastname: Bischoff
authors.forename: G
authors.initials: G
authors.lastname: Marko-Varga
mesh_terms: D002851:Chromatography, High Pressure Liquid / Q000379:methods*; D002852:Chromatography, Ion Exchange / Q000379:methods*; D010449:Peptide Mapping; D011506:Proteins / Q000737:chemistry*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity
chemical_list: D011506:Proteins