In vitro validation of volumetric blood flow measurement using Doppler flow wire.
Determination of any volumetric blood flow requires assessment of mean blood flow velocity and vessel cross-sectional area. For evaluation of coronary blood flow and flow reserve, however, assessment of average peak velocity alone is widely used, but changes in velocity profile and vessel area are not taken into account. We studied the feasibility of a new method for calculation of volumetric blood flow by Doppler power using a Doppler flow wire. An in vitro model with serially connected silicone tubes of known lumen diameters (1.5, 2.0, 2.5, 3.0, 3.5 and 4.0 mm) and pulsatile blood flow ranging from 10 to 200 mL/min was used. A Doppler flow wire was connected to a commercially available Doppler system (FloMap(R), Cardiometrics) for online calculation of the zeroth (M(0)) and the first (M(1)) Doppler moment, as well as mean flow velocity (V(m)). Two different groups of sample volumes (at different gate depths) were used: 1. two proximal sample volumes lying completely within the vessel were required to evaluate the effect of scattering and attenuation on Doppler power, and 2. distal sample volumes intersecting completely the vessel lumen to assess the vessel cross-sectional area. Area (using M(0)) and V(m) (using M(1)/M(0)) obtained from the distal gates were corrected for scattering and attenuation by the data obtained from the proximal gates, allowing calculation of absolute volumetric flow. These results were compared to the respective time collected flow. Correlation between time collected and Doppler-derived flow measurements was 0.98 (p < 0.0001), with a regression line close to the line of equality indicating an excellent agreement of the two measurements in each individual tube. The mean paired flow difference between the two techniques was 1.5 +/- 9.0 mL/min (ns). Direct volumetric blood flow measurement from received Doppler power using a Doppler flow wire system is feasible. This technique may potentially be of great clinical value because it allows an accurate assessment of coronary flow and flow reserve with a commercially available flow wire system.
Jenni
R
R
University Hospital Zurich, Division of Cardiology, Echocardiography, Zurich, Switzerland.
Kaufmann
P A
PA
Jiang
Z
Z
Attenhofer
C
C
Linka
A
A
Mandinov
L
L
D001783:Blood Flow Velocity; D001810:Blood Volume; D003326:Coronary Circulation; D005240:Feasibility Studies; D006801:Humans; D008955:Models, Cardiovascular; D019047:Phantoms, Imaging; D011673:Pulsatile Flow; D018608:Ultrasonography, Doppler / Q000295:instrumentation*