title: Energy correction factors of LiF powder TLDs irradiated in high-energy electron beams and applied to mailed dosimetry for quality assurance networks.
abstract: Absorbed dose determination with thermoluminescent dosimeters (TLDs) generally relies on calibration in 60Co gamma-ray reference beams. The energy correction factor fCo(E) for electron beams takes into account the difference between the response of the TLD in the beam of energy E and in the 60Co gamma-ray beam. In this work, fCo(E) was evaluated for an LiF powder irradiated in electron beams of 6 to 20 MeV (Varian 2300C/D) and 10 to 50 MeV (Racetrack MM50), and its variation with electron energy, TLD size and nature of the surrounding medium was also studied for LiF powder. The results have been applied to the ESTRO-EQUAL mailed dosimetry quality assurance network. Monte Carlo calculations (EGS4, PENELOPE) and experiments have been performed for the LiF powder (rho = 1.4 g cm3) (DTL937, Philitech, France), read on a home made reader and a PCL3 automatic reader (Fimel, France). The TLDs were calibrated using Fricke dosimetry and compared with three ionization chambers (NE2571, NACP02, ROOS). The combined uncertainties in the experimental fCo(E) factors determined in this work are less than about 0.4% (1 SD), which is appreciably smaller than the uncertainties up to 1.4% (1 SD) reported for other calculated values in the literature. Concerning the Varian 2300C/D beams, the measured fCo(E) values decrease from 1.065 to 1.049 +/- 0.004 (1 SD) when the energy at depth in water increases from 2.6 to 14.1 MeV; the agreement with Monte Carlo calculations is better than 0.5%. For the Racetrack MM50 pulsed-scanned beams, the average experimental value of fCo(E) is 1.071 +/- 0.005 (1 SD) for a mean electron energy at depth Ez ranging from 4.3 to 36.3 MeV: fCo(E) is up to 2% higher for the MM50 beams than for the 2300C/D beams in the range of the tested energies. The energy correction factor for LiF powder (3 mm diameter and 15 mm length) varies with beam quality and type (pulsed or pulsed-scanning), cavity size and nature of the surrounding medium. The fCo(E) values obtained for the LiF powder (3 mm diameter and 15 mm length) irradiated in water, have been applied to the EQUAL external audit network, leading to a good agreement between stated and measured doses, with a mean value of 1.002 +/- 0.022 (1 SD), for 170 beam outputs checked (36 electron beam energies) in 13 'reference' radiotherapy centres in Europe. Such fCo(E) data improve the accuracy of the absorbed dose TLD determination in electron beams, justifying their use for quality control in radiotherapy.
authors.affiliation: Service de Physique, Institut Gustave-Roussy, Villejuif, France.
authors.forename: D
authors.initials: D
authors.lastname: Marre
authors.forename: I H
authors.initials: IH
authors.lastname: Ferreira
authors.forename: A
authors.initials: A
authors.lastname: Bridier
authors.forename: A
authors.initials: A
authors.lastname: Björeland
authors.forename: H
authors.initials: H
authors.lastname: Svensson
authors.forename: A
authors.initials: A
authors.lastname: Dutreix
authors.forename: J
authors.initials: J
authors.lastname: Chavaudra
mesh_terms: D002138:Calibration; D004583:Electrons; D005459:Fluorides / Q000737:chemistry; D018020:Lithium Compounds / Q000737:chemistry; D009010:Monte Carlo Method; D019904:Polymethyl Methacrylate / Q000737:chemistry; D011137:Polystyrenes / Q000737:chemistry; D011786:Quality Control; D011829:Radiation Dosage; D011874:Radiometry / Q000295:instrumentation* / Q000379:methods*; D015203:Reproducibility of Results; D012984:Software; D013696:Temperature; D014867:Water
chemical_list: D018020:Lithium Compounds; D011137:Polystyrenes; D014867:Water; C027651:lithium fluoride; D019904:Polymethyl Methacrylate; D005459:Fluorides