Analyte-receptor binding and dissociation kinetics for biosensor applications: a fractal analysis.
A fractal analysis of confirmative nature only is presented for analyte-receptor binding and dissociation kinetics for biosensor applications. Data taken from the literature may be modeled, in the case of binding using a single-fractal analysis or a dual-fractal analysis. The dual-fractal analysis represents a change in the binding mechanism as the reaction progresses on the surface. Relationships are presented for the binding and dissociation rate coefficients as a function of their corresponding fractal dimension, Df or the degree of heterogeneity that exists on the surface. When analyte-receptor binding or dissociation is involved, an increase in the heterogeneity on the surface (increase in Df) leads to an increase in the binding and in the dissociation rate coefficient. It is suggested that an increase in the degree of heterogeneity on the surface leads to an increase in the turbulence on the surface owing to the irregularities on the surface. This turbulence promotes mixing, minimizes diffusional limitations, and leads subsequently to an increase in the binding and in the dissociation rate coefficient (Martin S.J., Granstaff, V.E., Frye, G.C., Anal. Chem., 65, (1991) 2910). The binding and the dissociation rate coefficient are rather sensitive to the degree of heterogeneity, Df,bind and Df,diss respectively, that exists on the biosensor surface. For example, the order of dependence on Df,bind is 19.2 for the binding rate coefficient, kbind for the binding of 0.03-1.0 microM SH-2Ld in solution to 2C TCR immobilized on a surface plasmon resonance (SPR) biosensor (Corr, M., Salnetz, A.E., Boyd, L.F., Jelonek, M.T., Khilko, S., Al-Ramadi, B.K., Kim, Y.S., Maher, S.E., Bothwell, A.L.M., Margulies, D.H., Science, 265, (1994) 946). The order of dependence on Df,diss is -6.22 for the dissociation rate coefficient, kdiss for the dissociation of 250-1000 nM Sophora japonica agglutinin (SJA)-lactose complex from the SPR surface. In general, the technique is applicable to other reactions occurring on different types of surfaces, such as cell-surface reactions.
Ramakrishnan
A
A
Chemical Engineering Department, University of Mississippi, MS 38677-1848, USA.
Sadana
A
A
D015374:Biosensing Techniques; D017709:Fractals; D007700:Kinetics; D037102:Lectins / Q000378:metabolism*; D037121:Plant Lectins; D020349:Surface Plasmon Resonance
D037102:Lectins; D037121:Plant Lectins; C045476:sophora japonica agglutinin