title: Clean conditions for the determination of ultra-low levels of mercury in ice and snow samples.
abstract: Laboratory facilities and methods are presented for the determination of ultra-low levels of mercury (Hg) in ice and snow samples originating from polar ice caps or temperate regions. Special emphasis will be given to the presentation of the clean laboratory and the cleaning procedures. The laboratory is pressurized with air filtered through high efficiency particle filters. This first filtration is not enough to get rid of contamination by Hg in air. Experiments are conducted in a clean bench, especially built for Hg analysis, equipped with both particle filter and activated charcoal filter. It allows to obtain very low levels of atmospheric Hg contamination. Ultrapure water is produced for cleaning all the plastic containers that will be used for ice and snow samples and also for the dilution of the standards. Hg content in laboratory water is about 0.08+/-0.02 pg/g. A Teflon system has been developed for the determination of Hg in ice and snow samples based on Hg(II) reduction to Hg(0) with a SnCl2/HNO3 solution followed by the measurement of gaseous Hg(0) with a Hg analyzer GARDIS 1A+ based on the Cold Vapor Atomic Absorption Spectroscopy method. Blank determination is discussed.
authors.affiliation: Laboratoire de Glaciologie et Géophysique de l'Environnement du CNRS, associé à l'Université Joseph Fourier, Saint Martin D'Heres, France. ferrari@glaciog.ujf-grenoble.fr
authors.forename: C P
authors.initials: CP
authors.lastname: Ferrari
authors.forename: A L
authors.initials: AL
authors.lastname: Moreau
authors.forename: C F
authors.initials: CF
authors.lastname: Boutron