title: Test to evaluate countercurrent chromatographs. Liquid stationary phase retention and chromatographic resolution.
abstract: Countercurrent chromatography (CCC) is a liquid chromatography (LC) technique with a special column able to retain a liquid stationary phase while the liquid mobile phase is pumped through. The coil planet centrifuge machines are made of open tube wound on spools. A simple test is proposed. The methanol-water (90:10, v/v)-heptane biphasic system is used with heptane as the mobile phase in the ascending or tail-to-head mode. The methanol-water stationary phase retention volume is measured at different flow-rates and rotor rotation speeds. After every machine equilibration, an alkylbenzene mixture is injected and the retention factors, peak efficiencies and resolution factors are measured or calculated for each solute. The wealth of information contained in the data set obtained is demonstrated. Four coil planet centrifuge machines of very different characteristics and one hydrostatic CCC machine with channels and ducts were submitted to the test. It was shown that the Sf, stationary retention factor, obtained with these machines was linearly dependent on the square root of F, the mobile phase flow-rate [Q. Du, C. Wu, G. Qian, P. Wu, Y. Ito, J. Chromatogr. A 835 (1999) 231-235]. It is shown that the slopes of the Sf versus F(1/2) lines could be related to a minimum rotor rotation, omega(mini), necessary to obtain the hydrodynamic equilibrium. The Sf and F parameters give the mobile phase linear velocity, u. It is shown that u is proportional to the square root of omega, the rotor rotation speed. The slope and intercept of the latter relationship also result in an omega(mini) value coherent with the first one. With the peak efficiencies and chromatographic resolution factors obtained for toluene and hexylbenzene, the parameters: number of plates per tubing turn, machine volume for one plate, and tubing length for one plate, were calculated and compared for the five machines. The internal diameter of the tubing used is shown to be a critical parameter acting on the machine volume and number of tubing turns.
authors.affiliation: Laboratoire des Sciences Analytiques, CNRS, Universit de Lyon 1, Villeurbanne, France. berthod@univ-lyon1.fr
authors.forename: A
authors.initials: A
authors.lastname: Berthod
authors.forename: B
authors.initials: B
authors.lastname: Billardello
mesh_terms: D001554:Benzene / Q000032:analysis; D002498:Centrifugation; D002853:Chromatography, Liquid / Q000295:instrumentation / Q000379:methods*; D008433:Mathematics; D011786:Quality Control
chemical_list: D001554:Benzene