High-throughput, semi-automated determination of a cyclooxygenase II inhibitor in human plasma and urine using solid-phase extraction in the 96-well format and high-performance liquid chromatography with post-column photochemical derivatization-fluorescence detection.
Compound I, 5-chloro-3-(4-methanesulfonylphenyl)-6'-methyl-[2,3']bipyridinyl, has been found to be a specific inhibitor of the enzyme cyclooxygenase II (COX II). The anti-inflammatory properties of this compound are currently being investigated. HPLC assays for the determination of this analyte in human plasma and human urine have been developed. Isolation of I and the internal standard (II) was achieved by solid-phase extraction (SPE) in the 96-well format. A C8 SPE plate was used for the extraction of the drug from human plasma (recovery >90%) while a mixed-mode (C8/Cation) SPE plate was used to isolate the analytes from human urine (recovery approximately 71%). The analyte and internal standard were chromatographed on a Keystone Scientific Prism-RP guard column (20 x 4.6 mm) connected to a Prism-RP analytical column (150 x 4.6 mm), using a mobile phase consisting of 45% acetonitrile in 10 mM acetate buffer (pH = 4); the analytes eluted at retention times of 5.2 and 6.9 min for I and II, respectively. Compounds I and II were found to form highly fluorescent products after exposure to UV light (254 nm). Thus, the analytes were detected by fluorescence (lambda(ex) = 260 nm, lambda(em) =375 nm) following post-column photochemical derivatization. Eight point calibration curves over the concentration range of 5-500 ng/ml for human plasma and human urine yielded a linear response (R2>0.99) when a 1/y weighted linear regression model was employed. Based on the replicate analyses (n = 5) of spiked standards, the within-day precision for both assays was better than 7% C.V. at all points on the calibration curve; within-day accuracy was within 5% of nominal at all standard concentrations. The between-run precision and accuracy of the assays, as calculated from the results of the analysis of quality control samples, was better than 8% C.V. and within 8% of nominal. I was found to be stable in human plasma and urine for at least 8 and 2 months, respectively. In addition, the human plasma assay was semi-automated in order to improve sample throughput by utilizing a Packard liquid handling system and a Tom-Tec Quadra 96 SPE system. The precision and accuracy of the semi-automated procedure were comparable to the manual procedure. Over 5000 clinical samples have been analyzed successfully using these methods.
Merck Research Laboratories, Department of Drug Metabolism, West Point, PA 19486, USA.
C Z
CZ
Matthews
E J
EJ
Woolf
L
L
Lin
W
W
Fang
J
J
Hsieh
S
S
Ha
R
R
Simpson
B K
BK
Matuszewski
D001331:Automation; D002851:Chromatography, High Pressure Liquid / Q000379:methods*; D051546:Cyclooxygenase 2; D052246:Cyclooxygenase 2 Inhibitors; D016861:Cyclooxygenase Inhibitors / Q000097:blood / Q000493:pharmacokinetics* / Q000652:urine; D006801:Humans; D007527:Isoenzymes / Q000187:drug effects*; D008565:Membrane Proteins; D010777:Photochemistry; D011451:Prostaglandin-Endoperoxide Synthases / Q000187:drug effects*; D012015:Reference Standards; D012680:Sensitivity and Specificity; D013050:Spectrometry, Fluorescence / Q000379:methods*
D052246:Cyclooxygenase 2 Inhibitors; D016861:Cyclooxygenase Inhibitors; D007527:Isoenzymes; D008565:Membrane Proteins; D051546:Cyclooxygenase 2; C496540:PTGS2 protein, human; D011451:Prostaglandin-Endoperoxide Synthases