Cone-beam computed tomography with a flat-panel imager: magnitude and effects of x-ray scatter.
A system for cone-beam computed tomography (CBCT) based on a flat-panel imager (FPI) is used to examine the magnitude and effects of x-ray scatter in FPI-CBCT volume reconstructions. The system is being developed for application in image-guided therapies and has previously demonstrated spatial resolution and soft-tissue visibility comparable or superior to a conventional CT scanner under conditions of low x-ray scatter. For larger objects consistent with imaging of human anatomy (e.g., the pelvis) and for increased cone angle (i.e., larger volumetric reconstructions), however, the effects of x-ray scatter become significant. The magnitude of x-ray scatter with which the FPI-CBCT system must contend is quantified in terms of the scatter-to-primary energy fluence ratio (SPR) and scatter intensity profiles in the detector plane, each measured as a function of object size and cone angle. For large objects and cone angles (e.g., a pelvis imaged with a cone angle of 6 degrees), SPR in excess of 100% is observed. Associated with such levels of x-ray scatter are cup and streak artifacts as well as reduced accuracy in reconstruction values, quantified herein across a range of SPR consistent with the clinical setting. The effect of x-ray scatter on the contrast, noise, and contrast-to-noise ratio (CNR) in FPI-CBCT reconstructions was measured as a function of SPR and compared to predictions of a simple analytical model. The results quantify the degree to which elevated SPR degrades the CNR. For example, FPI-CBCT images of a breast-equivalent insert in water were degraded in CNR by nearly a factor of 2 for SPR ranging from approximately 2% to 120%. The analytical model for CNR provides a quantitative understanding of the relationship between CNR, dose, and spatial resolution and allows knowledgeable selection of the acquisition and reconstruction parameters that, for a given SPR, are required to restore the CNR to values achieved under conditions of low x-ray scatter. For example, for SPR = 100%, the CNR in FPI-CBCT images can be fully restored by: (1) increasing the dose by a factor of 4 (at full spatial resolution); (2) increasing dose and slice thickness by a factor of 2; or (3) increasing slice thickness by a factor of 4 (with no increase in dose). Other reconstruction parameters, such as transaxial resolution length and reconstruction filter, can be similarly adjusted to achieve CNR equal to that obtained in the scatter-free case.
Siewerdsen
J H
JH
Department of Radiation Oncology, William Beaumont Hospital, Royal Oak, Michigan 48073, USA. jsiewerd@beaumont.edu
Jaffray
D A
DA
D055592:Biophysical Phenomena; D001703:Biophysics; D006801:Humans; D019047:Phantoms, Imaging; D012542:Scattering, Radiation; D014057:Tomography, X-Ray Computed / Q000295:instrumentation* / Q000706:statistics & numerical data; D014867:Water; D014963:X-Ray Intensifying Screens
D014867:Water