High-throughput, semi-automated determination of a cyclooxygenase II inhibitor in human plasma and urine using solid-phase extraction in the 96-well format and high-performance liquid chromatography with post-column photochemical derivatization-fluorescence detection.
Compound I, 5-chloro-3-(4-methanesulfonylphenyl)-6'-methyl-[2,3']bipyridinyl, has been found to be a specific inhibitor of the enzyme cyclooxygenase II (COX II). The anti-inflammatory properties of this compound are currently being investigated. HPLC assays for the determination of this analyte in human plasma and human urine have been developed. Isolation of I and the internal standard (II) was achieved by solid-phase extraction (SPE) in the 96-well format. A C8 SPE plate was used for the extraction of the drug from human plasma (recovery >90%) while a mixed-mode (C8/Cation) SPE plate was used to isolate the analytes from human urine (recovery approximately 71%). The analyte and internal standard were chromatographed on a Keystone Scientific Prism-RP guard column (20 x 4.6 mm) connected to a Prism-RP analytical column (150 x 4.6 mm), using a mobile phase consisting of 45% acetonitrile in 10 mM acetate buffer (pH = 4); the analytes eluted at retention times of 5.2 and 6.9 min for I and II, respectively. Compounds I and II were found to form highly fluorescent products after exposure to UV light (254 nm). Thus, the analytes were detected by fluorescence (lambda(ex) = 260 nm, lambda(em) =375 nm) following post-column photochemical derivatization. Eight point calibration curves over the concentration range of 5-500 ng/ml for human plasma and human urine yielded a linear response (R2>0.99) when a 1/y weighted linear regression model was employed. Based on the replicate analyses (n = 5) of spiked standards, the within-day precision for both assays was better than 7% C.V. at all points on the calibration curve; within-day accuracy was within 5% of nominal at all standard concentrations. The between-run precision and accuracy of the assays, as calculated from the results of the analysis of quality control samples, was better than 8% C.V. and within 8% of nominal. I was found to be stable in human plasma and urine for at least 8 and 2 months, respectively. In addition, the human plasma assay was semi-automated in order to improve sample throughput by utilizing a Packard liquid handling system and a Tom-Tec Quadra 96 SPE system. The precision and accuracy of the semi-automated procedure were comparable to the manual procedure. Over 5000 clinical samples have been analyzed successfully using these methods.
Matthews
C Z
CZ
Merck Research Laboratories, Department of Drug Metabolism, West Point, PA 19486, USA.
Woolf
E J
EJ
Lin
L
L
Fang
W
W
Hsieh
J
J
Ha
S
S
Simpson
R
R
Matuszewski
B K
BK
D001331:Automation; D002851:Chromatography, High Pressure Liquid / Q000379:methods*; D051546:Cyclooxygenase 2; D052246:Cyclooxygenase 2 Inhibitors; D016861:Cyclooxygenase Inhibitors / Q000097:blood / Q000493:pharmacokinetics* / Q000652:urine; D006801:Humans; D007527:Isoenzymes / Q000187:drug effects*; D008565:Membrane Proteins; D010777:Photochemistry; D011451:Prostaglandin-Endoperoxide Synthases / Q000187:drug effects*; D012015:Reference Standards; D012680:Sensitivity and Specificity; D013050:Spectrometry, Fluorescence / Q000379:methods*
D052246:Cyclooxygenase 2 Inhibitors; D016861:Cyclooxygenase Inhibitors; D007527:Isoenzymes; D008565:Membrane Proteins; D051546:Cyclooxygenase 2; C496540:PTGS2 protein, human; D011451:Prostaglandin-Endoperoxide Synthases