Fundamental differences between micro- and nano-Raman spectroscopy.
Electric field polarization orientations and gradients close to near-field scanning optical microscope (NSOM) probes render nano-Raman fundamentally different from micro-Raman spectroscopy. With x-polarized light incident through an NSOM aperture, transmitted light has x, y and z components allowing nano-Raman investigators to probe a variety of polarization configurations. In addition, the strong field gradients in the near-field of a NSOM probe lead to a breakdown of the assumption of micro-Raman spectroscopy that the field is constant over molecular dimensions. Thus, for nano-Raman spectroscopy with an NSOM, selection rules allow for the detection of active modes with intensity dependent on the field gradient. These modes can have similar activity as infra-red absorption modes. The mechanism can also explain the origin and intensity of some Raman modes observed in surface enhanced Raman spectroscopy.
Ayars
E J
EJ
Physics Department, North Carolina State University, Raleigh, NC 27695-8202, USA.
Jahncke
C L
CL
Paesler
M A
MA
Hallen
H D
HD