title: [Comparative biomechanical compression trials with a new vertebral prosthetic implant].
abstract: The authors present a new titanium implant for replacement of the vertebral body (Synex). Possible indications would be fractures or dislocations with destruction of the anterior column, posttraumatic kyphosis as well as tumors in the throracolumbar spine. The construction has to be completed by a stabilizing implant. For best fit and contact to adjacent end-plates Synex is distractable in situ. The possibility of secondary dislocation or loss of correction should thereby be minimised.
  
  
  OBJECTIVE
  We performed comparative compression tests with Synex and MOSS ("Harms mesh cage") on human cadaveric specimens of intact vertebrae (L1). The aim of the study was to measure the compressive strength of the vertebral body end-plate in uniaxial loading via both implants to exclude a caving of Synex in vivo.
  
  
  METHODS
  12 human cadaveric specimens of intact vertebrae (L1) were divided in 2 similar groups (matched pairs) according to bone mineral density (BMD), determined using DE-QCT. The specimens were loaded with axial compression force at a constant speed of 5 mm/min to failure and the displacement was recorded with a continuous load-displacement curve.
  
  
  RESULTS
  The mean ultimate compression force (Fmax) showed a tendency towards a higher result testing Synex with 3396 N versus 2719 N (non significant). The displacement until Fmax was 2.9 mm in group S (Synex), which was half as long as in group M (5.8 mm). The difference was significant (p < 0.001). The compression force was twice as high and significantly (p < 0.05) higher with Synex at a displacement of 1 mm, 1.5 mm and 2 mm. A significant (p < 0.001) correlation (R = 0.89) between Fmax and BMD was found.
  
  
  CONCLUSIONS
  Synex was found to be at least comparable to MOSS for suspensory replacement of the vertebral body at the thoracolumbar spine. A possible consequence of the significantly higher mean compression forces between 1 and 2 mm displacement might be a decreased segmental deformation or loss of correction.
authors.affiliation: Unfallchirurgische Klinik, Medizinische Hochschule Hannover, Carl-Neuberg-Strasse 1, 30625 Hannover. Dr.Christian.Knop@t-online.de
authors.forename: C
authors.initials: C
authors.lastname: Knop
authors.forename: U
authors.initials: U
authors.lastname: Lange
authors.forename: L
authors.initials: L
authors.lastname: Bastian
authors.forename: M
authors.initials: M
authors.lastname: Oeser
authors.forename: M
authors.initials: M
authors.lastname: Blauth
mesh_terms: D000328:Adult; D001696:Biomechanical Phenomena; D019245:Compressive Strength; D005260:Female; D006801:Humans; D004204:Joint Dislocations / Q000601:surgery*; D007738:Kyphosis / Q000601:surgery*; D008159:Lumbar Vertebrae / Q000601:surgery; D008297:Male; D008422:Materials Testing; D008875:Middle Aged; D019736:Prostheses and Implants; D011474:Prosthesis Design; D016103:Spinal Fractures / Q000601:surgery*; D013124:Spinal Injuries / Q000601:surgery*; D013125:Spinal Neoplasms / Q000601:surgery*; D014025:Titanium
chemical_list: D014025:Titanium