title: Quality-control issues on high-resolution diagnostic monitors.
abstract: Previous literature indicates a need for more data collection in the area of quality control of high-resolution diagnostic monitors. Throughout acceptance testing, which began in June 2000, stability of monitor calibration was analyzed. Although image quality on all monitors was found to be acceptable upon initial acceptance testing using VeriLUM software by Image Smiths, Inc (Germantown, MD), it was determined to be unacceptable during the clinical phase of acceptance testing. High-resolution monitors were evaluated for quality assurance on a weekly basis from installation through acceptance testing and beyond. During clinical utilization determination (CUD), monitor calibration was identified as a problem and the manufacturer returned and recalibrated all workstations. From that time through final acceptance testing, high-resolution monitor calibration and monitor failure rate remained a problem. The monitor vendor then returned to the site to address these areas. Monitor defocus was still noticeable and calibration checks were increased to three times per week. White and black level drift on medium-resolution monitors had been attributed to raster size settings. Measurements of white and black level at several different size settings were taken to determine the effect of size on white and black level settings. Black level remained steady with size change. White level appeared to increase by 2.0 cd/m2 for every 0.1 inches decrease in horizontal raster size. This was determined not to be the cause of the observed brightness drift. Frequency of calibration/testing is an issue in a clinical environment. The increased frequency required at our site cannot be sustained. The medical physics division cannot provide dedicated personnel to conduct the quality-assurance testing on all monitors at this interval due to other physics commitments throughout the hospital. Monitor access is also an issue due to radiologists' need to read images. Some workstations are in use 7 AM to 11 PM daily. An appropriate monitor calibration frequency must be established during acceptance testing to ensure unacceptable drift is not masked by excessive calibration frequency. Standards for acceptable black level and white level drift also need to be determined. The monitor vendor and hospital staff agree that currently, very small printed text is an acceptable method of determining monitor blur, however, a better method of determining monitor blur is being pursued. Although monitors may show acceptable quality during initial acceptance testing, they need to show sustained quality during the clinical acceptance-testing phase. Defocus, black level, and white level are image quality concerns, which need to be evaluated during the clinical phase of acceptance testing. Image quality deficiencies can have a negative impact on patient care and raise serious medical-legal concerns. The attention to quality control required of the hospital staff needs to be realistic and not have a significant impact on radiology workflow.
authors.affiliation: Naval Medical Center, Portsmouth, VA, USA.
authors.forename: L F
authors.initials: LF
authors.lastname: Parr
authors.forename: A L
authors.initials: AL
authors.lastname: Anderson
authors.forename: B K
authors.initials: BK
authors.lastname: Glennon
authors.forename: P
authors.initials: P
authors.lastname: Fetherston
mesh_terms: D002138:Calibration; D016288:Computer Terminals / Q000592:standards*; D003626:Data Display; D006801:Humans; D011786:Quality Control; D011873:Radiology Information Systems; D012984:Software