A comparison of polyethylene wear in hips with cobalt-chrome or zirconia heads. A prospective, randomised study.
We have reviewed 70 patients with bilateral simultaneous total hip arthroplasties to determine the rate of failure and to compare polyethylene wear and osteolysis between an implant with a cobalt-chrome head and Hylamer liner with that of a zirconia head and Hylamer liner. The mean thickness of the polyethylene liner was 11.0 mm (8.8 to 12.2) in the hip with a zirconia head and 10.7 mm (8.8 to 12.2) in that with a cobalt-chrome head. At follow-up at 6.4 years no acetabular or femoral component had been revised for aseptic loosening and no acetabular or femoral component was loose according to radiological criteria in both the cemented and cementless groups. The mean rate of linear wear and annual wear rate were highest in the 22 mm zirconia femoral head (1.25 mm (SD 1.05) and 0.21 mm (SD 0.18), respectively) and lowest in the 22 mm cobalt-chrome femoral head (0.70 mm (SD 0.39) and 0.12 mm (SD 0.07), respectively). The mean volumetric wear was highest in the 28 mm zirconia femoral head (730.79 mm ) and lowest in the 22 mm cobalt-chrome femoral head (264.67 mm3), but if the results were compared by size of the femoral head and type of material there was no statistical difference (p > 0.05). Sequential measurements of annual wear showed that the zirconia femoral head had a relatively higher rate of penetration than the cobalt-chrome head over the first three years; thereafter the rate of wear was reduced and compared favourably with that of cobalt-chrome heads. There was a statistically significant relationship between the wear of the polyethylene liner and the age of the patient, male gender and the degree of abduction angle of the cup, but not diagnosis, weight, hip score, range of movement, or amount of anteversion. Osteolysis was identified on both sides of the acetabulum in six patients (9%). Of 12 hips with acetabular osteolysis, six had a 28 mm cobalt-chrome femoral head and the remaining six a 28 mm zirconia head. Osteolysis was observed in zones 1A and 7A of the femur in two hips (3%) with a 28 mm zirconia head (cemented hip) and in four (6%) with a 28 mm cobalt-chrome femoral head (cementless hip). Our findings suggest that although the performance of a zirconia femoral head with a Hylamer liner was not statistically different from that of a cobalt-chrome femoral head and Hylamer liner, there was a trend for the zirconia head to be worse than the cobalt-chrome femoral head.
Kim
Y H
YH
Joint Replacement Centre of Korea, Hae Min General Hospital, Seoul, Korea.
Kim
J S
JS
Cho
S H
SH
D000328:Adult; D002858:Chromium Alloys; D019544:Equipment Failure Analysis; D005260:Female; D005500:Follow-Up Studies; D006622:Hip Prosthesis; D006801:Humans; D008297:Male; D008875:Middle Aged; D010014:Osteolysis / Q000000981:diagnostic imaging / Q000601:surgery; D020959:Polyethylene; D011183:Postoperative Complications / Q000000981:diagnostic imaging / Q000601:surgery*; D011474:Prosthesis Design; D011859:Radiography; D012086:Reoperation; D012307:Risk Factors; D015040:Zirconium
D002858:Chromium Alloys; D020959:Polyethylene; D015040:Zirconium; C028541:zirconium oxide