Third ventriculostomy.
Since its description by Dandy in 1922, several techniques have been used to perform third ventriculostomy under endoscopic control. Except for the blunt technique, in which the endoscope is used by itself to create the opening in the floor of the third ventricle, the other techniques require more than one instrument to perforate the floor of the ventricle and enlarge the ventriculostomy. The new device described is a sterilizable modified forceps that allows both the opening of the floor and the enlargement of the ventriculostomy in a simple and effective way. The new device has the following characteristics: 1) the tip of the forceps is thin enough to allow the easy perforation of the floor of the ventricle; 2) the inner surface of the jaws is smooth to avoid catching vessels of the basal cistern; and 3) the outer surface of the jaws has indentations that catch the edges of the opening to prevent them from slipping along the instrument's jaws. The ventricle floor is opened by gentle pressure of the forceps, which is slowly opened so that the edges of the aperture are caught by the distal outer indentation of the jaws, leading to an approximately 4-mm opening of the floor. This device has been used successfully in 10 consecutive patients. This new device allows surgeons to perform third ventriculostomy under endoscopic control in a very simple, quick, and effective way, avoiding the need for additional single-use instruments.
Vandertop
W P
WP
van der Zwan
A
A
Verdaasdonk
R M
RM
D004724:Endoscopy; D004867:Equipment Design; D006801:Humans; D013525:Surgical Instruments; D020542:Third Ventricle / Q000601:surgery*; D014696:Ventriculostomy / Q000295:instrumentation*