title: Resistance of the lumbar spine against axial compression forces after implantation of three different posterior lumbar interbody cages.
abstract: BACKGROUND
  The aim of using interbody fusion cages is to distract the degeneratively decreased disc height to decompress the neural structures in the intervertebral foramina and allow bony fusion. Prerequisite for a successful fusion therapy is a high resistance against subsidence and breakage.
  
  
  METHODS
  Three types of implants, a cylindrical threaded titanium cage (Ray) (1c), a bullet shaped PEEK cage (Stryker) (1a) and a rectangular titanium cage with an endplate anchorage device (Marquardt) (1b) were implanted in eight monosegmental lumbar spine specimens (L 2/3 and L 4/5). Each specimen underwent a cyclic loading test with 40000 cycles at a rate of 5 Hz. A cyclic axial compression force ranging from 200 Newton [N] to 1000 N was applied and the axial translation recorded simultaneously to determine the subsidence tendency. After this procedure the specimens were tested with a progressive axial force until breakage.
  
  
  RESULTS
  There were only small differences in the subsidence tendency for the three cage designs. The height reduction due to cyclic loading ranged between 0.9 mm (Marquardt), 1.2 mm (Stryker) and 1.4 mm (Ray). The median break force ranged from 5486 N (Marquardt), 8359 N (Stryker) to 8413 N (Ray). No correlation between bone mineral density and failure load could be detected.
  
  
  CONCLUSIONS
  Endplate preparation and cage design of the tested implants do not seem to influence the resistance of the segment against cyclic axial compression. The compression with a continuously increasing load revealed that an implant-bone failure is not to be expected in physiological limits for all three cage types.
authors.affiliation: Department of Neurosurgery, Academic Hospital Munich-Bogenhausen, Technical University Munich, Munich, Germany.
authors.forename: M
authors.initials: M
authors.lastname: Krammer
authors.forename: R
authors.initials: R
authors.lastname: Dietl
authors.forename: C B
authors.initials: CB
authors.lastname: Lumenta
authors.forename: A
authors.initials: A
authors.lastname: Kettler
authors.forename: H J
authors.initials: HJ
authors.lastname: Wilke
authors.forename: A
authors.initials: A
authors.lastname: Büttner
authors.forename: L
authors.initials: L
authors.lastname: Claes
mesh_terms: D001696:Biomechanical Phenomena; D019245:Compressive Strength; D004867:Equipment Design; D006801:Humans; D008159:Lumbar Vertebrae / Q000473:pathology / Q000601:surgery*; D008422:Materials Testing; D019736:Prostheses and Implants; D013123:Spinal Fusion / Q000295:instrumentation* / Q000379:methods; D016474:Weight-Bearing