Air-cooled cold trap channel integrated in a microfluidic device for monitoring airborne BTEX with an improved detection limit.
We integrated an air-cooled cold trap (CT) channel in a microfluidic device for monitoring airborne benzene, toluene, ethylbenzene, and xylene (BTEX) gases and demonstrated its effect on improving the detection limit of the microfluidic device. The device consists of concentration and detection cells formed of 3 x 1 cm Pyrex plates. We first introduced a sample gas into the concentration cell, and the gas was adsorbed onto an adsorbent in the channel. We then raised the temperature using a thin-film heater and introduced the desorbed gas into the detection cell. To prevent dilution of the gas before detection, we propose an improvement to the concentration cell structure that involves the integration of the CT channel. We examined the CT effect by comparing three types of concentration cell with different channel structures. We found that we could detect a gas concentration about 2 orders of magnitude lower than in our previous work by optimizing the channel structure and integrating a CT channel. As an example of BTEX detection,we obtained a 0.05 ppm detection limit for toluene gas with a sampling time of 30 min.
Ueno
Yuko
Y
NTT Lifestyle and Environmental Technology Laboratories, Atsugi, Kanagawa, Japan.
Horiuchi
Tsutomu
T
Niwa
Osamu
O
D000327:Adsorption; D000393:Air Pollutants / Q000032:analysis*; D001554:Benzene / Q000032:analysis; D001555:Benzene Derivatives / Q000032:analysis; D006841:Hydrocarbons, Aromatic / Q000032:analysis*; D008832:Microchemistry / Q000295:instrumentation*; D012680:Sensitivity and Specificity; D013057:Spectrum Analysis; D014050:Toluene / Q000032:analysis; D014992:Xylenes / Q000032:analysis
D000393:Air Pollutants; D001555:Benzene Derivatives; D006841:Hydrocarbons, Aromatic; D014992:Xylenes; D014050:Toluene; D001554:Benzene; C004912:ethylbenzene