Implantable loop recorder undersensing mimicking complete heart block.
The newer implantable loop recorders recently introduced (Reveal Plus. Medtronic Inc, Minneapolis, MN, U.S.A.). equipped with auto activation capabilities, are expected to expand our diagnostic abilities in the investigation of syncopal episodes. The case presented here is that of a man with pre-syncopal and syncopal episodes, without organic heart disease, in whom the device was auto-activated due to undersensing and the recorded strip imitated complete heart block. Increased attention should be given by the interpreter of the device's recordings to avoid false positive diagnoses that might lead to serious therapeutic actions.
Cardiology Department, University Hospital of Heraklion, Crete, Greece.
S I
SI
Chrysostomakis
E N
EN
Simantirakis
M E
ME
Marketou
P E
PE
Vardas
D000368:Aged; D003951:Diagnostic Errors; D015716:Electrocardiography, Ambulatory / Q000295:instrumentation*; D004567:Electrodes, Implanted; D006327:Heart Block / Q000175:diagnosis*; D006801:Humans; D008297:Male; D013575:Syncope / Q000175:diagnosis* / Q000209:etiology / Q000503:physiopathology