[Intensity modulated radiotherapy (IMRT) with compensators].
The irradiation with intensity-modulated fields is possible with static as well as dynamic methods. In our university hospital, the intensity-modulated radiotherapy (IMRT) with compensators was prepared and used for the first time for patient irradiation in July 2001. The compensators consist of a mixture of tin granulate and wax, which is filled in a milled negative mould. The treatment planning is performed with Helax-TMS (MDS Nordion). An additional software is used for editing the modulation matrix ("Modifix"). Before irradiation of the first patient, extensive measurements have been carried out in terms of quality assurance of treatment planning and production of compensators. The results of the verification measurements have shown that IMRT with compensators possesses high spatial and dosimetric exactness. The calculated dose distributions are applied correctly. The accuracy of the calculated monitor units is normally better than 3%; in small volumes, further dosimetric inaccuracies between calculated and measured dose distributions are mostly less than 3%. Therefore, the compensators contribute to the achievement of high-level IMRT even when apparatuses without MLC are used. This paper describes the use of the IMRT with compensators, presents the limits of this technology, and discusses the first practical experiences.
Salz
Henning
H
Radiologische Klinik, Klinikum der Friedrich-Schiller-Universität Jena. henning.salz@med.uni-jena.de
Wiezorek
Tilo
T
Scheithauer
Marcel
M
Kleen
Werner
W
Schwedas
Michael
M
Wendt
Thomas G
TG
D004867:Equipment Design; D006801:Humans; D019047:Phantoms, Imaging; D011880:Radiotherapy Planning, Computer-Assisted / Q000295:instrumentation* / Q000379:methods