[Acute dysfunction of Liotta mitral bioprosthesis 8 years after implantation].
The low-profile porcine bioprosthesis Liotta was introduced in clinical practice in order to reduce the potential complications related to excessive protrusion of the stent into the left ventricle after mitral valve replacement. Otherwise the particular design of this device seems to determine an excessive stress at the commissural level. We describe a case of acute dysfunction of the Liotta bioprosthesis secondary to commissural tear associated with minimal calcifications that required a reoperation in emergency.
Ius
Paolo
P
U.O. di Cardiochirurgia, Dipartimento Cardiovascolare, Ospedale Regionale S. Maria dei Battuti 31100 Treviso. pius@ulss.tv.it
Biffis
Claudio
C
Valfré
Carlo
C
D000368:Aged; D001705:Bioprosthesis; D004452:Echocardiography; D004630:Emergencies; D005260:Female; D005500:Follow-Up Studies; D006350:Heart Valve Prosthesis; D006801:Humans; D008943:Mitral Valve; D011475:Prosthesis Failure; D012086:Reoperation; D013997:Time Factors