title: Non-radiation induced signals in TL dosimetry.
abstract: One source of background signals, which are non-radiation related, is the reader system and it includes dark current, external contaminants and electronic spikes. These factors can induce signals equivalent to several hundredths of mSv. Mostly, the effects are minimised by proper design of the TLD reader, but some effects are dependent on proper operation of the system. The other main group of background signals originates in the TL crystal and is due to tribothermoluminescence, dirt, chemical reactions and stimulation by visible or UV light. These factors can have a significant contribution, equivalent to over several mSv, depending on whether the crystal is bare or protected by PTFE. Working in clean environments, monitoring continuously the glow curves and performing glow curve deconvolution are suggested to minimise non-radiation induced spurious signals.
authors.affiliation: Nuclear Research Center Negev, Beer Sheva, Israel. gur118@inter.net.il
authors.forename: U
authors.initials: U
authors.lastname: German
authors.forename: M
authors.initials: M
authors.lastname: Weinstein
mesh_terms: D003460:Crystallization; D003624:Darkness; D004583:Electrons; D008027:Light; D008163:Luminescent Measurements; D011138:Polytetrafluoroethylene; D015203:Reproducibility of Results; D012621:Seasons; D013819:Thermoluminescent Dosimetry / Q000379:methods*
chemical_list: D011138:Polytetrafluoroethylene