On modelling the kinestatic charge detector for digital radiographic diagnostic and portal imaging.
The kinestatic charge detection (KCD) principle has been a digital radiography technique for more than a decade. The advances of the KCD technique have gone from diagnostic imaging to portal imaging. However, little work has been done on understanding the selection of key KCD parameters and relationships between them. In the present study, an engineering model was established that could be used to optimize the placements of key parameters in terms of KCD system mechanical design. In the proposed KCD engineering model, the basic energy conservation law was applied to the process of ion transmission. It allows for the computation of the KCD design parameters such as the optimum grid placement, high voltage board tilt angle and grid wire space, as well as to provide recommendations on high voltage board and electric potentials and their ratio.
Qi
G
G
Department of Mechanical Engineering, The University of Memphis, Memphis, Tennessee, USA.
Goloubev
M Y
MY
DiBianca
F A
FA
Samant
S
S
D003198:Computer Simulation; D017076:Computer-Aided Design; D004867:Equipment Design / Q000379:methods; D008962:Models, Theoretical; D011786:Quality Control; D011856:Radiographic Image Enhancement / Q000295:instrumentation* / Q000379:methods; D011874:Radiometry / Q000295:instrumentation* / Q000379:methods; D011880:Radiotherapy Planning, Computer-Assisted / Q000295:instrumentation* / Q000379:methods; D012680:Sensitivity and Specificity; D055672:Static Electricity; D014159:Transducers