A quality management program in intravascular brachytherapy.
While simple, intravascular brachytherapy (IVB) presents a considerable potential for harm to the patient. The medical physicist maintains the responsibility to minimize the likelihood of operational problems or dosimetric errors. The principals for safe operation remain the same as with any radiotherapy treatment: to deliver the correct dose, to the correct location, safety. To develop an effective and comprehensive quality management (QM) program for IVB, a physicist should utilize proven risk assessment techniques rather than simply thinking of things to check, and follow guidances such as ISO9001:2000. The proposed QM program includes the following: Procedures designed to assure the safety of the patient. Identification of the patient; tests of the integrity and patency for the delivery catheter, operation of the source train, and patency of the catheter in the treatment position; a check for recovery preparations; and verification of source recovery. Procedures to assure positional accuracy of the treatment: Verification of the positioning the catheter in the artery and of the sources in the catheter. Procedures to assure dosimetry accuracy: Acceptance testing of the device, including verification of the source strength and uniformity, and of the treatment duration tables; verification of the treatment prescription and duration for each patient; and control measures that minimize the likelihood of errors removing the source at the correct time.
Chakri
Abderrahim
A
Department of Medical Physics, 1530 Medical Science Center, University of Wisconsin-Madison, Madison, Wisconsin 53706, USA.
Thomadsen
Bruce
B
D001918:Brachytherapy / Q000295:instrumentation* / Q000379:methods*; D003327:Coronary Disease / Q000532:radiotherapy* / Q000628:therapy; D006801:Humans; D011785:Quality Assurance, Health Care; D011874:Radiometry / Q000379:methods; D011880:Radiotherapy Planning, Computer-Assisted