title: [Improved design of permanent maglev impeller assist heart].
abstract: Magnetic bearing has no mechanical contact between the rotor and stator. And a rotary pump with magnetic bearing has therefore no mechanical wear and thrombosis due to bearing. The available magnetic bearings, however, are devised with electric magnets, need complicated control and remarkable energy consumption. Resultantly, it is difficult to apply an electric magnetic bearing to rotary pump without disturbing its simplicity, implantability and reliability. The authors have developed a levitated impeller pump merely with permanent magnets. The rotor is supported by permanent magnetic forces radially. On one side of the rotor, the impeller is fixed; and on the other side of the rotor, the driven magnets are mounted. Opposite to this driven magnets, a driving motor coil with iron corn magnets is fastened to the motor axis. Thereafter, the motor drives the rotor via a rotating magnetic field. By laboratory tests with saline, if the rotor stands still or rotates under 4,000 rpm, the rotor has one-point contact axially with the driving motor coil. The contacting point is located in the center of the rotor. As the rotating speed increases gradually to more than 4,000 rpm, the rotor will detache from the stator axially. Then the rotor will be fully levitated. Since the axial levitation is produced by hydraulic force and the driven magnets have a gyro-effect, the rotor rotates very steadly during levitation. As a left ventricular assist device, the pump works in a rotating speed range of 5,000-8,000 rpm, the levitation of the impeller hence is ensured by practical use of the pump.
authors.affiliation: Institute of Biomedical Engineering, University of Jiangsu, Zhenjiang 212013.
authors.forename: Kunxi
authors.initials: K
authors.lastname: Qian
authors.forename: Pei
authors.initials: P
authors.lastname: Zeng
authors.forename: Weimin
authors.initials: W
authors.lastname: Ru
authors.forename: Haiyu
authors.initials: H
authors.lastname: Yuan
mesh_terms: D004867:Equipment Design; D006353:Heart-Assist Devices; D008280:Magnetics / Q000295:instrumentation*