Can virtual simulation of breast tangential portals accurately predict lung and heart volumes?
A treatment portal or simulator image has traditionally been used to demonstrate the lung and heart coverage of the breast tangential portal. In many cases, these images were acquired as a planning session on the linear accelerator. The patients were also CT scanned to assess the lung/heart volume and to determine the surgical site depth for the electron-boost energy. A study using 50 consecutive patients was performed comparing the digitally reconstructed radiograph (DRR) from the virtual simulation with treatment portal images. Modification to the patient's arm position is required when performing the planning CT scans due to the aperture size of the CT scanner. Virtual simulation was used to assess the potential variation of lung and heart measurements. The average difference in lung volume between the DRR and portal image was less than 2 mm, with a range of 0-5 mm. Arm position did not have a significant impact on field deviation; however, great care was taken to minimize any changes in arm position. The modification of the arm position for CT scanning did not lead to significant variations between the DRRs and portal images. The Advantage Sim software has proven capable of producing good quality DRR images, providing a realistic representation of the lung and heart volume included in the treatment portal.
Cooke
Stacey
S
Division of Oncology, Royal Brisbane Hospital, Brisbane, Queensland, Australia.
Rattray
Greg
G
D001943:Breast Neoplasms / Q000532:radiotherapy*; D003198:Computer Simulation; D006321:Heart / Q000000981:diagnostic imaging* / Q000528:radiation effects; D006801:Humans; D007091:Image Processing, Computer-Assisted; D008168:Lung / Q000000981:diagnostic imaging* / Q000528:radiation effects; D019047:Phantoms, Imaging; D011187:Posture; D011859:Radiography; D011880:Radiotherapy Planning, Computer-Assisted