title: The thrust plate prosthesis in patients with aseptic osteonecrosis of the femoral head.
abstract: BACKGROUND
  The thrust plate prosthesis (TPP) is an implant with a metaphyseal fixation at the proximal femur that transmits the load forces of the hip onto the femoral neck. The osseous incorporation of the TPP and the adaptation of the bone to this force transmission depend on the bone quality, which is reduced to minor vitality and stability in patients with osteonecrosis of the femoral head. Depending on the etiology of the femoral head necrosis, the TPP might lead to early failures.
  
  
  METHODS
  In a prospective study, 63 patients with 72 cementless TPP due to femoral head osteonecrosis were examined. A clinical and radiological evaluation was performed preoperatively, 3 and 6 months postoperatively, and every year thereafter. The average follow-up period was 4.8+/-1.3 years with a minimum of 3 years. The pathogenesis of femoral head necrosis included alcoholism (n=19), subsequent to renal transplantation (n=11), during cortisone therapy of other dyscrasia (n=9), preceding a polychemotherapy (n=4), diabetes (n=3), sickle cell anemia (n=1), and idiopathic osteonecrosis (n=25).
  
  
  RESULTS
  The Harris Hip Score increased continuously from 50.0 points beyond 79.8 points after 3 months to 86.8 points within the 1st year, and subsequently remained stable at this level. Revision was necessary in six cases (8.3%). Of these, three had an aseptic loosening of the implant: 2 cases with renal transplantation and 1 of alcoholism with an extension of the necrotic area to the seating of the TPP. The other three patients showed septic implant loosenings: 2 cases with renal transplantation and 1 of alcoholism. Radiolucent lines were found in 9 cases (12.5%), mostly in zones 1 and 2 underneath the TPP. Of these, 1 with an idiopathic osteonecrosis was assessed to be radiologically loosened. The overall failure rate was 9.7%, with a proportion of 36.4% in patients with renal transplantation. Excluding this specific patient group, the failure rate was 4.9%.
  
  
  CONCLUSIONS
  Femoral head necrosis following renal transplantation and extension of the necrotic area into the femoral neck are contraindications for TPP. Excluding these patients, the TPP shows comparable mid-term results to cementless stemmed prostheses and supplies advantages especially for younger patients, because of its metaphyseal, bone-preserving fixation. However, evaluation of the clinical impact of the TPP in comparison with other cementless femoral stem systems requires long-term examinations in the future.
authors.affiliation: Orthopaedic Department, University Hospital Hamburg-Eppendorf, Germany. fink@uke.uni-hamburg.de
authors.forename: Bernd
authors.initials: B
authors.lastname: Fink
authors.forename: Thomas
authors.initials: T
authors.lastname: Schneider
authors.forename: Silke
authors.initials: S
authors.lastname: Conrad
authors.forename: Marcus
authors.initials: M
authors.lastname: Jaeger
authors.forename: Michael
authors.initials: M
authors.lastname: Protzen
authors.forename: Wolfgang
authors.initials: W
authors.lastname: Rüther
mesh_terms: D000328:Adult; D019644:Arthroplasty, Replacement, Hip; D001696:Biomechanical Phenomena; D005260:Female; D005271:Femur Head Necrosis / Q000209:etiology / Q000601:surgery*; D006622:Hip Prosthesis; D006801:Humans; D008297:Male; D008875:Middle Aged; D011446:Prospective Studies; D012086:Reoperation