[Acceptance check and quality control of SPECT].
This paper explains the testing of SPECT, especially the new SPECT with double digital detector and spiral scanning frames that has been introduced to China recently, in the acceptance check, proceeding from the physical functions of the system to its mechanical functions, to the NEMA standard functions, and then to the computer hardware specified in the contract. A brief introduction is also given of the quality control of SPECT in terms of its spatial resolution, energy resolution, spatial linearity, sensitivity, and center of rotation.
Sun
L M
LM
Fujian Provincial Hospital.
Liu
C B
CB
D011786:Quality Control; D015899:Tomography, Emission-Computed, Single-Photon / Q000295:instrumentation / Q000592:standards*