Stability of electron-beam energy monitor for quality assurance of the electron-beam energy from radiotherapy accelerators.
Information on electron energy is important in planning radiation therapy using electrons. The Geske 3405 electron beam energy monitor (Geske monitor, PTW Nuclear Associates, Carle Place, NY, USA) is a device containing nine ionization chambers for checking the energy of the electron beams produced by radiotherapy accelerators. We wondered whether this might increase the likelihood of ionization chamber trouble. In spite of the importance of the stability of such a quality assurance (QA) device, there are no reports on the stability of values measured with a Geske monitor. The purpose of this paper was therefore to describe the stability of a Geske monitor. It was found that the largest coefficient of variation (CV) of the Geske monitor measurements was approximately 0.96% over a 21-week period. In conclusion, the stability of Geske monitor measurements of the energy of electron beams from a linear accelerator was excellent.
Chida
Koichi
K
Department of Radiological Technology, College of Medical Sciences, Tohoku University, Sendai 980-8575, Japan. chida@rad.cms.tohoku.ac.jp
Saito
Haruo
H
Takai
Yoshihiro
Y
Zuguchi
Masayuki
M
Mitsuya
Masatoshi
M
Sakakida
Hideharu
H
Kohzuki
Masahiro
M
Yamada
Shogo
S
D004583:Electrons; D006801:Humans; D010315:Particle Accelerators; D011785:Quality Assurance, Health Care; D011834:Radiation Monitoring / Q000295:instrumentation*; D011882:Radiotherapy, High-Energy / Q000295:instrumentation*