Relative movements between Kinemax Plus tibial inserts and the tibial base-plates.
Tests were performed on six large Kinemax Plus knee bearings (snap-fit design) to evaluate the amount of movement between 10- and 15-mm-thick tibial inserts and the tibial base plates. The knee bearings were tested up to 1 x 10(6) cycles on the Durham six-station knee wear simulator which subjected the bearings to similar motion and loading profiles that would be experienced by the natural knee during walking. Although passive internal/external (I/E) rotation was allowed, no active I/E rotation was applied. The movement of the tibial inserts was measured with dial gauges (accuracy +/-0.01 mm) before and after the bearings were tested on the simulator, when unloaded, and throughout the tests while the bearings were being dynamically loaded in the simulator. Movement occurred between the tibial insert and the tibial base plate after initial assembly due to the snap-fit mechanism used to locate the tibial insert within the tibial base plate. However this decreased appreciably when the bearings were loaded in the simulator. The amount of movement did not change with time when the bearings were continuously loaded in the simulator. However, after each test the amount of movement of the tibial inserts, when unloaded, was only 65 per cent (anterior-posterior) and 46 per cent (medial-lateral) of the values before the test. This was thought to be due to creep of the ultra-high molecular weight polyethylene (UHMWPE) inserts. The movement between the tibial insert and tibial base plate in situ is likely to be much less than that observed by a surgeon at the time of assembly due to loading of the knee bearing in the body. However, the amount of movement when the tibial inserts are loaded may still be great enough to produce a second interface where wear of the tibial insert may take place.
Ash
H E
HE
Center for Biomedical Engineering, University of Durham, Durham DH1 3LE, UK.
Scholes
S C
SC
Parkin
R
R
Unsworth
A
A
D020099:Coated Materials, Biocompatible; D019544:Equipment Failure Analysis / Q000295:instrumentation / Q000379:methods*; D006801:Humans; D007719:Knee Joint / Q000503:physiopathology* / Q000601:surgery; D007720:Knee Prosthesis; D008153:Lubrication; D008422:Materials Testing / Q000295:instrumentation / Q000379:methods*; D009068:Movement; D011095:Polyethylenes; D011474:Prosthesis Design; D016059:Range of Motion, Articular; D012680:Sensitivity and Specificity; D013499:Surface Properties; D013977:Tibia / Q000503:physiopathology; D016138:Walking / Q000502:physiology; D016474:Weight-Bearing
D020099:Coated Materials, Biocompatible; D011095:Polyethylenes; C111601:ultra-high molecular weight polyethylene