title: Fixation of the offset V osteotomy: mechanical testing of 4 constructs.
abstract: The purpose of this study was to assess 4 methods of fixation for the offset V osteotomy. Maximum load, failure energy, and stiffness were determined in a control group and in 4 different test models. There were 10 specimens for each group. The control group consisted of intact first ray sawbones. An offset V osteotomy was performed on each specimen in each of the 4 test groups. The osteotomies were fixated either with two 2.0-mm cortical screws, two 2.7-mm cortical screws, two 3.5-mm cortical screws, or one 2.7-mm cortical screw and a 0.045-in Kirschner wire, respectively. Each model was then loaded to failure with a computerized hydraulic-tensile testing machine. Results for maximum load to failure for all 4 fixation constructs showed a mean ranging from 113.0 to 144.0 N, a mean energy to failure ranging from 272.2 to 365.0 J, and a mean stiffness ranging from 21.3 to 27.0 N/mm. There were no statistically significant differences detected among the individual constructs. In the group fixated with 3.5-mm screws, there were statistically significant differences in all 3 parameters compared with the control group. In the groups fixated with 2.7-mm screws, 2.0-mm screws, and the 2.7-mm screw and Kirschner wire, there were statistically significant differences in the maximum load and the failure energy, but not in stiffness, when compared with the control group. These findings suggest that all 4 constructs provide similar mechanical properties when utilized for the fixation of the offset V osteotomy.
authors.affiliation: St. John North Shores Hospital, Harrison Township, MI, and Michigan State University, East Lansing, MI, USA.
authors.forename: Jane
authors.initials: J
authors.lastname: Khuri
authors.forename: Stuart
authors.initials: S
authors.lastname: Wertheimer
authors.forename: Jill
authors.initials: J
authors.lastname: Krueger
authors.forename: Roger
authors.initials: R
authors.lastname: Haut
mesh_terms: D001696:Biomechanical Phenomena; D001863:Bone Screws / Q000592:standards*; D001864:Bone Wires / Q000592:standards*; D004868:Equipment Failure; D008682:Metatarsal Bones / Q000601:surgery; D008953:Models, Anatomic; D010027:Osteotomy / Q000295:instrumentation*