[Advanced method for calculating the scatter signal contribution in CT detectors by the Monte Carlo method].
Multislice spiral CT scanners allow to acquire multiple slices simultaneously. With increasing numbers of slices, not only the total extent of slice collimation increases, but also the contribution of scatter radiation to the detector signal. A fast method for calculating the scatter signal would offer the possibility to correct the measured detector signal. Monte Carlo methods allow to simulate the paths of photons through a 3D volume, both in a patient- and scanner-specific fashion. If a scatter photon leaves the volume, its path can be followed and its interaction with an element of the detector be checked. This conventional way of calculating the scatter signal is time-consuming. In order to reduce the calculation time, a more efficient method was developed (Method of Weights). Every time an interaction occurs inside of the 3D volume, the probability of a detector hit due to photon scattering is calculated for each detector channel. The respective value is added to the scatter signal per detector with the corresponding weight. Simulated values of scatter-to-primary-signal ratios were confirmed by data available in the literature. Both the conventional and fast methods for the calculation of scatter signals yielded identical values within the range of statistical accuracy. Assuming the same computing time, the standard deviation for the conventional method was 5 times higher than for the fast one. The presented method allows to significantly reduce the computation time. It may therefore provide a basis for "real time" methods to correct for the scatter signal, especially in case of increasing numbers of slices.
Institut für Medizinische Physik, Universität Erlangen-Nürnberg.
Bernhard
B
Schmidt
Willi A
WA
Kalender
D006801:Humans; D009010:Monte Carlo Method; D019047:Phantoms, Imaging; D015203:Reproducibility of Results; D012542:Scattering, Radiation; D014057:Tomography, X-Ray Computed / Q000379:methods*