Evaluation of the Vitek 2 ID-GNB assay for identification of members of the family Enterobacteriaceae and other nonenteric gram-negative bacilli and comparison with the Vitek GNI+ card.
We evaluated the Vitek 2 ID-GNB identification card (bioMérieux, Inc., Durham, N.C.) for its ability to identify members of the family Enterobacteriaceae and other gram-negative bacilli that are isolated in clinical microbiology laboratories. Using 482 enteric stock cultures and 103 strains of oxidase-positive, gram-negative glucose-fermenting and nonfermenting bacilli that were maintained at -70 degrees C and passaged three times before use, we inoculated cards according to the manufacturer's directions and processed them in a Vitek 2 instrument using version VT2-R02.03 software. All panel identifications were compared to reference identifications previously confirmed by conventional tube biochemical assays. At the end of the initial 3-h incubation period, the Vitek 2 instrument demonstrated an accuracy of 93.0% for the identification of enteric strains; 414 (85.9%) were correctly identified at probability levels ranging from excellent to good, and an additional 34 (7.1%) strains were correctly identified but at a low level of discrimination. Nineteen (3.9%) strains were unidentified, and 15 (3.1%) were misidentified. The 19 unidentified strains were scattered among 10 genera. Three of the 15 misidentified strains were lactose-positive Salmonella spp. and were identified as Escherichia coli; another was a lactose-positive, malonate-negative Salmonella enterica subsp. arizonae strain that was identified as E. coli. Of the 103 glucose-fermenting and nonfermenting nonenteric strains, 88 (85.4%) were correctly identified at probability levels ranging from excellent to good, and 10 (9.7%) were correctly identified, but at a low level of discrimination, for a total of 95.1% accuracy with this group. Two strains were unidentified and three were misidentified. The errors occurred for strains in three different genera. With the increased hands-off approach of the Vitek 2 instrument and accuracies of 93% for the identification of enteric organisms and 95.1% for the identification of nonenteric organisms with the ID-GNB card, use of this product presents an acceptable method for the identification of most gram-negative organisms commonly isolated in the clinical laboratory. A comparison of these results to those obtained by testing 454 of the same strains with the Vitek GNI+ card revealed no significant difference in the abilities of the two cards to identify these organisms accurately.
O'Hara
Caroline M
CM
Epidemiology and Laboratory Branch, Division of Healthcare Quality Promotion, Centers for Disease Control and Prevention, Atlanta, Georgia 30333, USA. cmo1@cdc.gov
Miller
J Michael
JM
D015373:Bacterial Typing Techniques / Q000295:instrumentation* / Q000706:statistics & numerical data; D004755:Enterobacteriaceae / Q000145:classification* / Q000302:isolation & purification / Q000378:metabolism; D005285:Fermentation; D005947:Glucose / Q000378:metabolism; D006090:Gram-Negative Bacteria / Q000145:classification* / Q000302:isolation & purification / Q000378:metabolism; D006801:Humans; D010088:Oxidoreductases / Q000378:metabolism; D013045:Species Specificity
D010088:Oxidoreductases; D005947:Glucose