Automatic continuous river monitoring of nitrate using a novel ion-selective electrode.
On-site continuous measurement of nitrate in the River Taw situated in Devon (UK) was performed automatically for two months using a recently developed ion-selective electrode (ISE). A non-submersible electrode support system was developed and a field instrument was constructed comprising the nitrate-ISEs, a reference electrode and a temperature probe connected through a pre-amplifier to a data-logger and battery supply. The nitrate-ISE was constructed using a commercially available electrode body with a membrane incorporated into the tip. This rubbery membrane contained the sensor molecule (N,N,N-triallylleucine betaine chloride, 6.5% m/m) covalently bound to polystyrene-block-polybutadiene-block-polystyrene, SBS, (43.5% m/m), with 2-nitrophenyloctyl ether (2-NPOE) as mediator (40% m/m). The Nernstian slope was -59.1 mV per decade over a linear range of 1400-0.07 mg L(-1) nitrate-N. The limit of detection was 0.007 mg L(-1) nitrate-N with a selectivity coefficient for nitrate against chloride (k(pot)NO3-, Cl-) of 3.4 x 10(-3). The speed of response was less than a minute over the linear Nernstian range. The lifetime in the laboratory exceeded 5 months with no potentiometric drift over the linear Nernstian range. A temperature correction algorithm was also used to accomodate the temperature changes encountered in the river. The river nitrate results obtained with the ISEs at hourly intervals compared very favourably (R2 = 0.9) with those obtained with laboratory automated chemical determinations made on contemporaneous river samples over a concentration range 0.55-2.00 mg L(-1) nitrate-N. The ISEs did not require re-calibration and no deterioration in performance or fouling of the membrane surface was observed during 40 days.
Plymouth Environment Research Centre, School of Environmental Sciences, University of Plymouth, Drake Circus, Plymouth, UK PL4 8AA.
Thierry
T
Le Goff
Jim
J
Braven
Les
L
Ebdon
David
D
Scholefield
D001331:Automation; D004566:Electrodes; D004784:Environmental Monitoring / Q000379:methods*; D004867:Equipment Design; D007477:Ions / Q000032:analysis; D009566:Nitrates / Q000032:analysis*; D014873:Water Pollutants / Q000032:analysis*
D007477:Ions; D009566:Nitrates; D014873:Water Pollutants