Evaluation of NO(x) flue gas analyzers for accuracy and their applicability for low-concentration measurements.
The requirements of the Texas State Implementation Plan of the U.S. Clean Air Act for the Houston-Galveston Ozone Nonattainment Area stipulate large reductions in oxides of nitrogen (NO(x)) emissions. A large number of sources at Dow Chemical Co. sites within the nonattainment area may require the addition of continuous emission monitoring systems (CEMS) for online analysis of NO(x), CO, and O2. At the outset of this work, it was not known whether the analyzers could accurately measure NO(x) as low as 2 ppm. Therefore, NO(x) CEMS analyzers from five different companies were evaluated for their ability to reliably measure NO(x) in the 2-20 ppm range. Testing was performed with a laboratory apparatus that accurately simulated different mixtures of flue gas and, on a limited basis, simulated a dual-train sampling system on a gas turbine. The results indicate that this method is a reasonable approach for analyzer testing and reveal important technical performance aspects for accurate NO(x) measurements. Several commercial analyzers, if installed in a CEMS application with sampling conditioning components similar to those used in this study, can meet the U.S. Environmental Protection Agency's measurement data quality requirements for accuracy.
Gluck
Steven
S
Dow Chemical Co., Freeport, Texas 77541, USA. sgluck@dow.com
Glenn
Chuck
C
Logan
Tim
T
Vu
Bac
B
Walsh
Mike
M
Williams
Pat
P
D000395:Air Pollutants, Occupational / Q000032:analysis*; D000397:Air Pollution / Q000331:legislation & jurisprudence*; D000465:Algorithms; D004784:Environmental Monitoring; D008163:Luminescent Measurements; D009566:Nitrates / Q000032:analysis*; D015203:Reproducibility of Results; D013056:Spectrophotometry, Ultraviolet; D013781:Texas
D000395:Air Pollutants, Occupational; D009566:Nitrates