Miniaturized movable contactless conductivity detection cell for capillary electrophoresis.
A miniaturized capacitively coupled contactless conductivity detector (mini-C(4)D) cell has been designed which is small enough to allow it to slide along the effective capillary length inside the capillary cassette of an Agilent capiillary electrophoresis system (CE) (or other CE brand of similar construction), including the possibility of positioning it close to the point of optical detection (4 cm), or even putting two such detector cells in one cassette. The cell was tested and the performance characteristics (noise, sensitivity, and peak width) were compared with those obtained with the previously used large C(4)D cell. No significant differences were observed. The mini-C(4)D was used in simultaneous separations of common cations and anions where its advantage over a larger C(4)D cell is the ability to vary the point of detection with the mini-C(4)D cell continuously at any point along the capillary length, so that the optimum apparent selectivity can be chosen. Other applications include providing a convenient second point of detection in addition to photometric detection, such as to measure accurately the linear velocity of a zone, or to allow placement of two mini-C(4)D cells in one capillary cassette simultaneously.
Macka
Miroslav
M
Australian Centre for Research on Separation Science, School of Chemistry, Faculty of Science, Engineering and Technology, University of Tasmania, Hobart, Australia.
Hutchinson
Joseph
J
Australian Centre for Research on Separation Science, School of Chemistry, Faculty of Science, Engineering and Technology, University of Tasmania, Hobart, Australia.
Zemann
Andreas
A
Institute of Analytical Chemistry and Radiochemistry, Leopold-Franzens-University of Innsbruck, Innsbruck, Austria.
Shusheng
Zhang
Z
Australian Centre for Research on Separation Science, School of Chemistry, Faculty of Science, Engineering and Technology, University of Tasmania, Hobart, Australia.
Haddad
Paul R
PR
Australian Centre for Research on Separation Science, School of Chemistry, Faculty of Science, Engineering and Technology, University of Tasmania, Hobart, Australia.
D004563:Electrochemistry / Q000295:instrumentation*; D019075:Electrophoresis, Capillary / Q000295:instrumentation*; D005069:Evaluation Studies as Topic; D008904:Miniaturization