Sampling atmospheric carbonaceous aerosols using an integrated organic gas and particle sampler.
Measurement of particle-bound organic carbon (OC) may be complicated by sampling artifacts such as adsorption of gas-phase species onto particles or filters or evaporation of semivolatile compounds off the particles. A denuder-based integrated organic gas and particle sampler (IOGAPS), specifically designed to minimize sampling artifacts, has been developed to sample atmospheric carbonaceous aerosols. IOGAPS is designed to first remove gas-phase chemicals via sorption to the XAD-coated denuder, and subsequently particles are trapped on a quartz filter. A backup sorbent system consisting of sorbent- (XAD-4 resin) impregnated filters (SIFs) was used to capture the semivolatile OC that evaporates from the particles accumulated on the upstream quartz filter. A traditional filter pack (FP) air sampler, which uses a single quartz filter to collect the particles, was employed for comparison in this study. Elemental and organic carbon were determined from filter punches by a thermal optical transmittance aerosol carbon analyzer. Field measurements show that there was no significant difference between the elemental carbon concentrations determined by the FP and IOGAPS, indicating that particle loss during the transit through the denuder tube was negligible. Compared with the OC determined by FP (3.9-12.6 microg of C/m3), the lower OC observed on the quartz filter in the IOGAPS (2.2-6.0 microg of C/m3) was expected because of the removal of gas-phase organics by the denuder. Higher semivolatile organic carbon (SVOC) on the backup SIFs during the night (1.24-8.43 microg of C/m3) suggests that more SVOC, emitted from primary sources or formed as secondary organic compounds, partitions onto the particles during the night because of the decreased ambient temperature. These data illustrate the utility of an IOGAPS system to more accurately determine the particle-bound OC in comparison to FP-based systems.
Fan
Xinghua
X
Department of Chemistry, University of Toronto, 80 St. George Street, Toronto, Ontario M5S 3H6, Canada.
Brook
Jeffrey R
JR
Mabury
Scott A
SA
D000336:Aerosols / Q000032:analysis*; D000393:Air Pollutants / Q000032:analysis*; D002244:Carbon / Q000032:analysis*; D004784:Environmental Monitoring / Q000295:instrumentation* / Q000379:methods*; D005374:Filtration; D055095:Optics and Photonics; D010316:Particle Size; D013696:Temperature; D014835:Volatilization
D000336:Aerosols; D000393:Air Pollutants; D002244:Carbon