title: [Comparison of shielding calculations for diagnostic X-ray equipment].
abstract: Radiation is controlled by the recent revision of the Enforcement Regulations of the Medical Service Law. Law No. 188 prescribes X-ray data for estimating effective dose in the controlled area of a medical facility. For X-ray data such as kerma in air, transmission data are based on NCRP. We compared various X-ray data, and NCRP and Simpkin were compared with law No. 188. Leakage effective doses in a general-purpose radiography room, fluoroscopy room, and X-ray CT room were calculated. All three calculations were below the dose limit for controlled areas, 1.3 mSv/3 months. In the general-purpose radiography room (including fluoroscopy), NCRP and Simpkin underestimated from 1/3 to 1/2 in comparison with law No. 188. In the X-ray CT room, NCRP and Simpkin showed underestimates from 1/2 to 2/3 compared with law No. 188. There were no significant differences between law No. 188, NCRP, and Simpkin, but significant differences were found for individual numerical values. A need for the re-examination of basic data was suggested by the above findings.
authors.affiliation: Graduate School of Medical Science, Kanazawa University.
authors.forename: Kimiya
authors.initials: K
authors.lastname: Noto
authors.forename: Takumi
authors.initials: T
authors.lastname: Sota
authors.forename: Kichiro
authors.initials: K
authors.lastname: Koshida
authors.forename: Shouichi
authors.initials: S
authors.lastname: Suzuki
mesh_terms: D005160:Facility Design and Construction; D011829:Radiation Dosage; D011835:Radiation Protection / Q000331:legislation & jurisprudence / Q000592:standards; D011859:Radiography / Q000295:instrumentation*; D011872:Radiology Department, Hospital; D011874:Radiometry; D012542:Scattering, Radiation