Implantation of a single-piece, hydrophilic, acrylic, minus-power foldable posterior chamber intraocular lens in a rabbit model: clinicopathologic study of posterior capsule opacification.
OBJECTIVE
To compare the extent of posterior capsule opacification (PCO) after implantation of a standard-power biconvex Centerflex intraocular lens (IOL) and a newly introduced biconcave high-minus-power Centerflex design in rabbit eyes.


METHODS
The Center for Research on Ocular Therapeutics and Biodevices, Storm Eye Institute, Medical University of South Carolina, Charleston, South Carolina, and the David J. Apple, MD, Laboratories for Ophthalmic Devices Research, John A. Moran Eye Center, University of Utah School of Medicine, Salt Lake City, Utah, USA.


METHODS
Twelve rabbits had phacoemulsification and implantation of 2 foldable single-piece hydrophilic acrylic Centerflex posterior chamber IOLs. The right eyes received a standard-power (+21.00 diopters [D]) biconvex-optic lens and the left eyes, a minus-power (-7.00 D) biconcave-optic IOL. Formation of PCO was evaluated 3 weeks after surgery using the Miyake-Apple posterior photography technique. Histological sections from each globe were prepared to analyze capsular bag status and assess postsurgical intracapsular lens epithelial cell (LEC) proliferation, especially ingrowth of LECs across the visual axis. The data were analyzed using the Kruskal-Wallis 1-way analysis of variance for nonparametric measurements and the Mann-Whitney rank sum test.


RESULTS
There was no significant difference in Soemmering's ring formation between the 2 IOL models. The biconcave minus-power IOL showed significantly lower central and peripheral PCO scores than the biconvex standard-power lens (P<.05). Pathological evaluations revealed that the effective site of blockage of LECs was at the truncated optic edge of both lenses, even in the presence of retained and/or regenerative cortical material.


CONCLUSIONS
This study confirms the efficacy of a truncated IOL optic in helping reduce the incidence of PCO. Both IOL designs have optic geometries that create clear-cut barrier effects. However, the biconcave minus-power IOL, which has a thicker, square, truncated optic edge with a ridge that encircles the periphery of the optic for 360 degrees, appears to have an enhanced barrier effect, especially at the optic-haptic junction. This further minimizes the ingrowth of migrating LECs toward the visual axis.
Vargas
Luis G
LG
David J Apple, MD Laboratories for Ophthalmic Devices Research, John A Moran Eye Center, University of Utah, Salt Lake City, Utah 84132, USA.
Izak
Andrea M
AM
Apple
David J
DJ
Werner
Liliana
L
Pandey
Suresh K
SK
Trivedi
Rupal H
RH
D000180:Acrylic Resins; D000818:Animals; D019923:Capsulorhexis; D002386:Cataract / Q000209:etiology / Q000473:pathology / Q000517:prevention & control*; D002455:Cell Division; D004847:Epithelial Cells / Q000473:pathology; D007903:Lens Capsule, Crystalline / Q000473:pathology*; D019654:Lens Implantation, Intraocular; D007910:Lenses, Intraocular; D023421:Models, Animal; D018918:Phacoemulsification / Q000009:adverse effects; D011183:Postoperative Complications / Q000473:pathology / Q000517:prevention & control*; D011474:Prosthesis Design; D011817:Rabbits
D000180:Acrylic Resins