title: Measurement of serum albumin by capillary zone electrophoresis, bromocresol green, bromocresol purple, and immunoassay methods.
abstract: OBJECTIVE
  The introduction of capillary zone electrophoresis (CZE) to this laboratory has highlighted discrepancies in albumin measured on an Abbott Aeroset by bromocresol green (BCG) and that calculated by CZE on the basis of total protein measured by Biuret.
  
  
  METHODS
  This study examined differences in albumin estimation by CZE, Abbott Aeroset BCG, and Aeroset bromocresol purple (BCP), and compared these with albumin estimated by Beckman Array immunoassay.
  
  
  RESULTS
  Altman and Bland analysis of results showed a positive bias of BCG with CZE (4.51 g/litre; 95% limits of agreement, 3.77 to 5.26; n = 72) and BCP (3.85 g/litre; 95% limits of agreement, -1.42 to 9.12; n = 72). CZE and BCP agreed closely (0.67 g/litre; 95% limits of agreement, -4.39 to 3.06; n = 72). Analysis of 57 of those samples in which BCG and CZE differed > or = 5 g/litre showed a positive bias of BCG with immunoassay (8.35 g/litre; 95% limits of agreement, 1.54 to 15.16; n =57), with good agreement between CZE and immunoassay (-0.44 g/litre; 95% limits of agreement, -2.82 to 1.94; n = 57).
  
  
  CONCLUSIONS
  BCP is superior to BCG for the assay of albumin and has replaced BCG as the routine test for albumin in this laboratory.
mesh_terms: D001961:Bromcresol Green; D001962:Bromcresol Purple; D019075:Electrophoresis, Capillary / Q000379:methods; D006801:Humans; D007118:Immunoassay / Q000379:methods; D012680:Sensitivity and Specificity; D012709:Serum Albumin / Q000032:analysis*; D013194:Staining and Labeling
chemical_list: D012709:Serum Albumin; D001962:Bromcresol Purple; D001961:Bromcresol Green