title: A canine model for studying endoleak after endovascular aneurysm repair.
abstract: OBJECTIVE
  The aim of this study was to create an animal model of endoleak after stent-graft placement for abdominal aortic aneurysm (AAA) in which a large aneurysmal sac would be preserved for the testing of techniques for its percutaneous occlusion.
  
  
  METHODS
  Infrarenal AAAs were created in nine dogs by anastomosis of an isolated segment of the inferior vena cava to the right side of the abdominal aorta in combination with a large anterior patch from the external jugular vein. One hour later, animals underwent percutaneous implantation of polytetrafluoroethylene-covered Z stent endografts with three 3-mm-diameter holes through the fabric. Aortograms were obtained before and after surgery, after endograft placement, and at the time of animal sacrifice at 1 week or 1, 2, 3, or 6 months. Pressures within the aorta and the aneurysm sac were recorded before animal sacrifice. Gross and histologic evaluations of the specimens were then carried out.
  
  
  RESULTS
  Immediately after endograft placement, all nine animals had artificial type III endoleaks with angiographic filling of lumbar arteries and veins. One animal died of surgical complications within 2 days of surgery and is not included in our data analysis. One aneurysm ruptured at 1 week. At completion of the study, six endografts were patent and two were occluded. The aneurysm sac had enlarged by approximately 50% in seven animals. At follow-up, type I endoleak was present in three animals, type II endoleak was present in three, and the artificial type III endoleak was present in all six animals with patent endografts. The pressure differential between aorta and aneurysm sac was 36 mm Hg, with a mean aortic pressure of 87 mm Hg +/- 13.3 and a mean aneurysmal sac pressure of 51 mm Hg +/- 28.1. The aneurysmal sac exhibited early thrombus formation at 1 week, which progressed to complete thrombosis in 1-6 months.
  
  
  CONCLUSIONS
  The model is technically feasible but would be useful in testing occlusive techniques for residual aneurysm sacs only in the acute phase after endograft placement. It would be not reliable for chronic evaluation because of rapidly progressive thrombosis in most aneurysm sacs and occasional complete thrombosis of the AAA and endograft.
authors.affiliation: Dotter Interventional Institute, Oregon Health and Science University, L342, 3181 SW Sam Jackson Park Road, Portland, Oregon 97201, USA. pavcnikd@ohsu.edu
authors.forename: Dusan
authors.initials: D
authors.lastname: Pavcnik
authors.forename: Robert T
authors.initials: RT
authors.lastname: Andrews
authors.forename: Qiang
authors.initials: Q
authors.lastname: Yin
authors.forename: Barry T
authors.initials: BT
authors.lastname: Uchida
authors.forename: Hans A
authors.initials: HA
authors.lastname: Timmermans
authors.forename: Christopher
authors.initials: C
authors.lastname: Corless
authors.forename: Naoyuki
authors.initials: N
authors.lastname: Toyota
authors.forename: Manabu
authors.initials: M
authors.lastname: Nakata
authors.forename: John
authors.initials: J
authors.lastname: Kaufman
authors.forename: Frederick S
authors.initials: FS
authors.lastname: Keller
authors.forename: Josef
authors.initials: J
authors.lastname: Rösch
mesh_terms: D000818:Animals; D017544:Aortic Aneurysm, Abdominal / Q000000981:diagnostic imaging / Q000473:pathology / Q000601:surgery*; D019917:Blood Vessel Prosthesis Implantation; D004285:Dogs; D011138:Polytetrafluoroethylene; D011183:Postoperative Complications / Q000000981:diagnostic imaging / Q000628:therapy; D011859:Radiography; D015607:Stents
chemical_list: D011138:Polytetrafluoroethylene