Comparison of methods for the measurement of mist and vapor from light mineral oil-based metalworking fluids.
The measurement of oil mist derived from metalworking fluids formulated with light mineral oils can be highly inaccurate when using traditional filter sampling. This is due to evaporation of oil from the filter. In this work the practicability of an alternative approach measuring total oil mist and vapor was investigated. Combinations of inhalable particle samplers with backup sorbent vapor traps and standard vapor sampling on pumped and diffusive sorbent tubes were evaluated with gravimetric, infrared spectroscopic, and gas chromatographic analytical methods against the performance requirements of European Standard EN 482. An artificial aerosol was used to compare the methods against a reference method of filter sampler in series with three impingers. Multi-orifice samplers were used with standard 8-mm diameter charcoal tubes at 2 L/min without any signs of channelling or significant breakthrough, as were conical inhalable samplers with XAD-2 tubes at 1 L/min. Most combinations of samplers had a bias of less than 3 percent, but solitary pumped charcoal tubes underestimated total oil by 13 percent. Diffusive sampling was affected by impaction of mist particles and condensation of oil vapor. Gravimetric analysis of filters revealed significant potential sample loss during storage, with 4 percent being lost after one day when stored at room temperature and 2 percent when refrigerated. Samples left overnight in the balance room to equilibrate lost 24 percent. Infrared spectroscopy gave more precise results for vapor than gas chromatography (p = 0.002). Gas chromatography was less susceptible to bias from contaminating solvent vapors than infrared spectroscopy, but was still vulnerable to petroleum distillates. Under the specific test conditions (one oil type and mist particle size), all combinations of methods examined complied with the requirements of European Standard EN 484. Total airborne oil can be measured accurately; however, care must be taken to avoid contamination by hydrocarbon solvent vapors during sampling.
Simpson
Andrew T
AT
Health and Safety Laboratory, Sheffield, United Kingdom.
D000336:Aerosols / Q000032:analysis; D000395:Air Pollutants, Occupational / Q000032:analysis; D004784:Environmental Monitoring / Q000295:instrumentation / Q000379:methods*; D007219:Industrial Oils / Q000032:analysis*; D008669:Metallurgy / Q000295:instrumentation*; D008899:Mineral Oil / Q000032:analysis*; D016273:Occupational Exposure / Q000032:analysis; D017550:Spectroscopy, Fourier Transform Infrared / Q000379:methods; D014835:Volatilization
D000336:Aerosols; D000395:Air Pollutants, Occupational; D008899:Mineral Oil