[The Sorin pericardial bioprosthesis--a stentless aortic valve with very good hemodynamic performance].
BACKGROUND
The Sorin Pericarbon Freedom Stentless aortic valve has the potential to provide superior hemodynamic function and durability. In this study we assessed the hemodynamic performance of this valve and its impact on LV-mass regression after aortic valve replacement.


METHODS
31 consecutive patients who received a Sorin Pericarbon Freedom Stentless aortic valve were analysed. Mean age of the patients (16 female and 15 male) was 64 +/- 17 years. Five patients had isolated aortic stenosis, three isolated aortic regurgitation and one combined aortic valve disease. Three patients had combined aortic and mitral valve disease, 14 patients concomitant coronary artery disease, one congenital aortic coarctation. Three patients had an acute aortic valve endocarditis. 18 patients were classified as high-risk patients (mean EuroSCORE 9 +/- 2). Mean left ventricular ejection fraction was 52.5 +/- 15.0%.


RESULTS
Valve sizes from 21 mm to 29 mm were implanted. The valves were oversized by 2 mm compared to measurement. 16 patients received isolated aortic valve replacement, Three patients aortic valve replacement and mitral valve reconstruction. 12 patients had concomitant CABG. Three procedures were reoperations. Hospital mortality was 6.4% (two patients). Both deaths occurred in high-risk patients and were not valve-related. Four patients had perioperative low-output-syndrome and needed IABP. After six months a follow up echocardiography was performed. Mean and peak gradients were 9.6 +/- 4.4 and 20.6 +/- 5.9 mmHg, respectively. Significant reduction of left ventricular hypertrophy (LV mass index 126.5 +/- 27.3 vs. 189.6 +/- 45.3 g/m2, p = 0.0313) and improvement of the ejection fraction (58 +/- 9.8 vs. 52.5 +/- 15.0%, p = 0.9749) as compared with preoperative valve.


CONCLUSIONS
The hemodynamic performance of the Sorin Pericarbon Freedom Stentless aortic valve is excellent and the patient outcome is satisfying. However the implantation technique requires longer ischemic time, the prosthesis offers very satisfying hemodynamic function and accelerates probably the LV-mass regression in the mid term follow-up. Late performance and durability of the valve have to be assessed.
Klinik für Herz- und Gefässchirurgie, Universitätsspital Inselspital, Bern, Schweiz.
V A
VA
Gegouskov
F S
FS
Eckstein
B
B
Kipfer
P A
PA
Berdat
F F
FF
Immer
J
J
Schmidli
C
C
Seiler
C
C
Zobrist
T P
TP
Carrel
D000368:Aged; D000369:Aged, 80 and over; D001022:Aortic Valve Insufficiency / Q000401:mortality / Q000503:physiopathology / Q000601:surgery*; D001024:Aortic Valve Stenosis / Q000401:mortality / Q000503:physiopathology / Q000601:surgery*; D001705:Bioprosthesis; D015897:Comorbidity; D004697:Endocarditis, Bacterial / Q000401:mortality / Q000503:physiopathology / Q000601:surgery; D005260:Female; D005500:Follow-Up Studies; D006350:Heart Valve Prosthesis; D006439:Hemodynamics / Q000502:physiology*; D006801:Humans; D008297:Male; D008875:Middle Aged; D011183:Postoperative Complications / Q000401:mortality / Q000503:physiopathology*; D011474:Prosthesis Design; D012307:Risk Factors; D015607:Stents; D015996:Survival Rate; D013557:Switzerland; D020257:Ventricular Remodeling / Q000502:physiology