On the determination of the post-irradiation time from the glow curve of TLD-100.
The ratio of peak 3 to the sum of peaks 4 + 5 in TLD-100 was measured for various pre-irradiation and post-irradiation time periods, under conditions characteristic of routine personal dosimetry. It was confirmed that the value of this ratio depends only on the elapsed time between the prior readout and the present one, independent of the moment when the irradiation took place during the total time interval (storage time). This effect indicates that fading of peak 3 seems to be due mainly to changes in the unoccupied traps, and not to decay of trapped charges, being almost independent of the presence of electrons or holes in the traps. This observation leads to the conclusion that the suggestions in the past to use the decay of peak 3 in TLD-100 for the measurement of the elapsed time between irradiation and readout may have been wrong. On the other hand, the decay of peak 2 can be used to measure the elapsed time from irradiation, since the rate of decay is different when related to pre-irradiation and post-irradiation times, indicating a much higher decay rate of the trapped charges (Randall-Wilkins decay). However, because of the fast decay rate of peak 2, its use for determination of the elapsed time since irradiation is of little practical significance.
Weinstein
M
M
Nuclear Research Centre-Negev, Beer Sheva, 84190, Israel.
German
U
U
Dubinsky
S
S
Alfassi
Z B
ZB
D019544:Equipment Failure Analysis / Q000379:methods*; D006207:Half-Life; D018499:Linear Energy Transfer; D008422:Materials Testing / Q000379:methods*; D011829:Radiation Dosage; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013819:Thermoluminescent Dosimetry / Q000295:instrumentation* / Q000379:methods*; D013997:Time Factors