Immunoaffinity column cleanup with liquid chromatography using post-column bromination for determination of aflatoxin B1 in cattle feed: collaborative study.
A collaborative study was conducted to evaluate the effectiveness of an immunoaffinity column cleanup liquid chromatography (LC) method for determination of aflatoxin B1 in cattle feed at a possible future European regulatory limit (1 ng/g). The test portion was extracted with acetone-water (85 + 15), filtered, diluted with water, and applied to an immunoaffinity column. The column was washed with water to remove interfering compounds, and the purified aflatoxin B1 was eluted with methanol. Aflatoxin B1 was separated and determined by reversed-phase liquid chromatography (RP-LC) and detected by fluorescence after post column derivatization (PCD) involving bromination. PCD was achieved with either pyridinium hydrobromide perbromide (PBPB), used by 14 laboratories, or an electrochemical cell and addition of bromide to the mobile phase, used by 7 laboratories. Both derivatization techniques were not significantly different when compared by the t-test; the method was statistically evaluated for all laboratories together (bromination and PBPB). The cattle feed samples, both spiked and naturally contaminated with aflatoxin B1, were sent to 21 laboratories in 14 different countries (United States, Japan, and Europe). Test portions were spiked at levels of 1.2 and 3.6 ng/g for aflatoxin B1. Recoveries ranged from 74 to 157%. Based on results for spiked samples (blind pairs at 2 levels) as well as naturally contaminated samples (blind pairs at 3 levels), the relative standard deviation for repeatability (RSDr) ranged from 5.9 to 8.7%. The relative standard deviation for reproducibility (RSDR) ranged from 17.5 to 19.6%. The method showed acceptable within- and between-laboratory precision for this matrix, as evidenced by HORRAT values, at the target levels of determination for aflatoxin B1. No major differences in RSD were observed, showing that the composition of the feeds was not a factor for the samples tested and that the method was applicable for all materials used.
European Commission, Directorate General Joint Research Centre, Institute for Reference Materials and Measurements, Retieseweg, B-2440 Geel, Belgium.
Joerg
J
Stroka
Christoph
C
von Holst
Elke
E
Anklam
Matthias
M
Reutter
D016604:Aflatoxin B1 / Q000032:analysis*; D000821:Animal Feed / Q000032:analysis*; D000818:Animals; D002138:Calibration; D002417:Cattle; D002846:Chromatography, Affinity; D007120:Immunochemistry; D007202:Indicators and Reagents; D013058:Mass Spectrometry; D015203:Reproducibility of Results; D013050:Spectrometry, Fluorescence
D007202:Indicators and Reagents; D016604:Aflatoxin B1