Application of pulse mixing method in software coincidence counting.
In recent years the software coincidence counting system, designed for absolute activity measurement, has been developed in the Czech Metrology Institute. In this system a true coincidence count rate is calculated from the records of time and amplitude data of individual pulses and may be determined by two different methods. The first one uses a coincidence resolving time, in a manner similar to a classical coincidence measurement. The second method applies the pulse mixing method formulae, so that it does not use the resolving time and the correction for accidental coincidences. Both methods have been tested on the same data from a 4pibeta (PC)-gamma coincidence measurement of 60Co sources. The difference between the results obtained from both calculation methods applied on the data from the same measurement did not exceed 0.015%. The details of both methods and the results of their comparison are presented.
Havelka
Miroslav
M
Czech Metrology Institute, Inspectorate for Ionizing Radiation, Radiová 1, Prague 102 00, Czech Republic. mhavelka@cmi.cz
Auerbach
Pavel
P
Sochorová
Jana
J
D000465:Algorithms; D001610:Beta Particles; D005720:Gamma Rays; D011874:Radiometry / Q000379:methods*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D012815:Signal Processing, Computer-Assisted; D012984:Software; D012986:Software Validation