Integration of single cell injection, cell lysis, separation and detection of intracellular constituents on a microfluidic chip.
A microfluidic system was developed for the analysis of single biological cells, with functional integration of cell sampling, single cell loading, docking, lysing, and capillary electrophoretic (CE) separation with laser induced fluorescence (LIF) detection in microfabricated channels of a single glass chip. Channels were 12 microm deep and 48 microm wide, with a simple crossed-channel design. The effective separation channel length was 35 mm. During sampling with a cell suspension (cell population 1.2 x 10(5) cells per mL in physiological salt solution), differential hydrostatic pressure (created by adjusting liquid levels in the four reservoirs) was used to control cell flow exclusively through the channel crossing. Single cell loading into the separation channel was achieved by electrophoretic means by applying a set of potentials at the four reservoirs, counteracting the hydrostatic flow. A special docking (adhering) procedure for the loaded cell was applied before lysis by repeatedly connecting and disconnecting a set of low potentials, allowing precise positioning of the cell within the separation channel. Cell lysis was then effected within 40 ms under an applied CE separation voltage of 1.4 kV (280 V cm(-1)) within the working electrolyte (pH 9.2 borate buffer) without additional lysates. The docked lysing approach reduced dispersion of released intracellular constituents, and significantly improved the reproducibility of CE separations. Glutathione (GSH) was used as a model intracellular component in single human erythrocyte cells. NDA derivatized GSH was detected using LIF. A throughput of 15 samples h(-1), a retention time precision of 2.4% RSD was obtained for 14 consecutively injected cells. The average cellular concentration of GSH in human erythrocytes was found to be 7.2 [times] 10(-4)+/- 3.3 x 10(-4) M (63 +/- 29 amol per cell). The average separation efficiency for GSH in lysed cells was 2.13 x 10(6)+/- 0.4 x 10(6) plates per m, and was about a factor of 5 higher than those obtained with GSH standards using pinched injection.
Gao
Jian
J
Institute of Microanalytical Systems, Department of Chemistry, Zhejiang University, 310028 Hangzhou, China.
Yin
Xue-Feng
XF
Fang
Zhao-Lun
ZL
D019075:Electrophoresis, Capillary / Q000295:instrumentation / Q000379:methods*; D004912:Erythrocytes / Q000737:chemistry; D005978:Glutathione / Q000032:analysis; D006801:Humans; D007424:Intracellular Fluid / Q000737:chemistry; D044085:Microfluidics; D009281:Naphthalenes; D013194:Staining and Labeling; D013997:Time Factors
D009281:Naphthalenes; C027717:2,3-naphthalenedicarboxaldehyde; D005978:Glutathione