Dosimetry characterization of 32P intravascular brachytherapy source wires using Monte Carlo codes PENELOPE and GEANT4.
Monte Carlo calculations using the codes PENELOPE and GEANT4 have been performed to characterize the dosimetric parameters of the new 20 mm long catheter-based 32P beta source manufactured by the Guidant Corporation. The dose distribution along the transverse axis and the two-dimensional dose rate table have been calculated. Also, the dose rate at the reference point, the radial dose function, and the anisotropy function were evaluated according to the adapted TG-60 formalism for cylindrical sources. PENELOPE and GEANT4 codes were first verified against previous results corresponding to the old 27 mm Guidant 32P beta source. The dose rate at the reference point for the unsheathed 27 mm source in water was calculated to be 0.215 +/- 0.001 cGy s(-1) mCi(-1), for PENELOPE, and 0.2312 +/- 0.0008 cGy s(-1) mCi(-1), for GEANT4. For the unsheathed 20 mm source, these values were 0.2908 +/- 0.0009 cGy s(-1) mCi(-1) and 0.311 0.001 cGy s(-1) mCi(-1), respectively. Also, a comparison with the limited data available on this new source is shown. We found non-negligible differences between the results obtained with PENELOPE and GEANT4.
Torres
Javier
J
Servicio de Radiofísica y Protección Radiológia, Hospital Universitario Virgen de la Arrixaca, E-30120 Murcia, Spain.
Buades
Manuel J
MJ
Almansa
Julio F
JF
Guerrero
Rafael
R
Lallena
Antonio M
AM
D016880:Anisotropy; D001918:Brachytherapy / Q000295:instrumentation* / Q000379:methods*; D002404:Catheterization; D004307:Dose-Response Relationship, Radiation; D015233:Models, Statistical; D009010:Monte Carlo Method; D019047:Phantoms, Imaging; D011874:Radiometry / Q000379:methods*; D012984:Software; D014867:Water
D014867:Water