title: Batteries: from alkaline to zinc-air.
abstract: There is no perfect disposable battery--one that will sit on the shelf for 20 years, then continually provide unlimited current, at a completely constant voltage until exhausted, without producing heat. There is no perfect rechargeable battery--one with all of the above characteristics and will also withstand an infinite overcharge while providing an equally infinite cycle life. There are only compromises. Every battery selection is a compromise between the ideally required characteristics, the advantages, and the limitations of each battery type. General selection of a battery type to power a medical device is largely outside the purview of the biomed. Initially, these are engineering decisions made at the time of medical equipment design and are intended to be followed in perpetuity. However, since newer cell types evolve and the manufacturer's literature is fixed at the time of printing, some intelligent substitutions may be made as long as the biomed understands the characteristics of both the recommended cell and the replacement cell. For example, when the manufacturer recommends alkaline, it is usually because of the almost constant voltage it produces under the devices' design load. Over time, other battery types may be developed that will meet the intent of the manufacturer, at a lower cost, providing longer operational life, at a lower environmental cost, or with a combination of these advantages. In the Obstetrical Doppler cited at the beginning of this article, the user had put in carbon-zinc cells, and the biomed had unknowingly replaced them with carbonzinc cells. If the alkaline cells recommended by the manufacturer had been used, there would have been the proper output voltage at the battery terminals when the [table: see text] cells were at their half-life. Instead, the device refused to operate since the battery voltage was below presumed design voltage. While battery-type substitutions may be easily and relatively successfully made in disposable applications (for example, zinc-air for alkaline--if it is cost-effective), this is absolutely forbidden for secondary cells. Because of the differing cell voltages, charge characteristics and overcharge tolerance between different types of secondary cells, substituting a nickel-cadmium battery pack for the more expensive lithium-ion pack (if it is physically able to fit into the battery compartment), might appear to save money (e.g. $50 vs. $100) but it would be very ill advised. Since the cell characteristics are very different, it would be downright fatal to anyone within the 'kill radius' when the pack explodes. Those outside the kill radius would receive chemical burns from the electrolyte. Substitutions of secondary cell battery packs are generally not a good idea for biomeds to engage in. These are engineering decisions best left to either aftermarket battery pack manufacturers or the medical device manufacturer as a design engineering change.
authors.affiliation: U.S. Military Entrance Processing Command, Chicago, IL, USA.
authors.forename: Robert M
authors.initials: RM
authors.lastname: Dondelinger
mesh_terms: D000388:Air; D000468:Alkalies; D011211:Electric Power Supplies / Q000145:classification*; D004868:Equipment Failure; D004864:Equipment and Supplies; D008670:Metals; D014481:United States
chemical_list: D000468:Alkalies; D008670:Metals