[Method of estimating patient skin dose from dose displayed on medical X-ray equipment with flat panel detector].
The International Electrotechnical Commission has stipulated that medical X-ray equipment for interventional procedures must display radiation doses such as air kerma in free air at the interventional reference point and dose area product to establish radiation safety for patients (IEC 60601-2-43). However, it is necessary to estimate entrance skin dose for the patient from air kerma for an accurate risk assessment of radiation skin injury. To estimate entrance skin dose from displayed air kerma in free air at the interventional reference point, it is necessary to consider effective energy, the ratio of the mass-energy absorption coefficient for skin and air, and the backscatter factor. In addition, since automatic exposure control is installed in medical X-ray equipment with flat panel detectors, it is necessary to know the characteristics of control to estimate exposure dose. In order to calculate entrance skin dose under various conditions, we investigated clinical parameters such as tube voltage, tube current, pulse width, additional filter, and focal spot size, as functions of patient body size. We also measured the effective energy of X-ray exposure for the patient as a function of clinical parameter settings. We found that the conversion factor from air kerma in free air to entrance skin dose is about 1.4 for protection.
Fukuda
Atsushi
A
Department of Radiology, Shiga Medical Center for Adults.
Koshida
Kichiro
K
Yamaguchi
Ichiro
I
Togashi
Atsuhiko
A
Matsubara
Kousuke
K
D000179:Acrylates; D006801:Humans; D015641:Radiography, Interventional / Q000295:instrumentation; D011874:Radiometry / Q000295:instrumentation* / Q000379:methods*; D012867:Skin / Q000528:radiation effects*
D000179:Acrylates