[Intra-prosthetic dislocation of the Bousquet dual mobility socket].
OBJECTIVE
The Bousquet system is a dual mobility head-polyethylene polyethylene-metal cup socket. The polyethylene insert retaining the femoral head moves in the noncemented metal cup, increasing both mobility and stability.


METHODS
Between 1989 and 1997, seven cases of intra-prosthetic dislocation (six patients) were observed. The femoral head escaped from the polyethylene insert due to wear. On the average, this complication occurred ten Years after implantation. Risk of dislocation was high in six of the seven hips. All patients had a large sized stem screwed into the femoral neck. There was a characteristic radiological aspect with loss of the concentric head metal cup configuration. The head was applied against the upper wall of the metal cup.


RESULTS
Surgical replacement was undertaken early in six patients by simply changing the insert without modifying the other stable components. Outcome remained good at three to eight Years. One patient underwent late surgery. The insert and the cup were replaced with a classical implant. Functional outcome was good but recurrent dislocation occurred.


CONCLUSIONS
At mid-term, intra-prosthetic dislocation of dual mobility sockets appears to be exceptional. Dislocation results from polyethylene wear leading to failure of the insert to retain the prosthetic head. Wear is favored by direct phenomena (direct contact between neck and insert which can occur early if there is a small difference in the head and neck diameters) or indirect phenomena (factors limiting polyethylene metal-cup mobility). Surgical treatment is necessary. If undertaken early, replacement with a modular head and insert can be sufficient if the prosthesis has not loosened but the metal cup may have to be replaced in the event of metal-metal contact between the head and the cup. Prosthesis loosening, wear of the metal cup, or an identified cause of dislocation imply replacing the failing implants. Implantation of the dual mobility system is particularly interesting for patients with a high risk of dislocation or a chronically unstable hip prosthesis. Careful technique is required to reduce or retard the risk of intra-prosthetic dislocation.


CONCLUSIONS
Intra-prosthetic dislocation of a dual mobility socket is an exceptional complication at mid-term. Surgical treatment is required but may be limited to simple insert replacement. Systematic use of this type of implant in young subjects must be carefully examined, but for us, the risk of dislocation does not outweigh the advantages of this original concept of dual mobility. This type of socket remains an useful preventive technique for high-risk hips or for curative treatment of recurrent dislocation.
Lecuire
F
F
Service de Chirurgie Orthopédique, Hôpital Renée-Sabran, boulevard Edouard-Herriot, 83406 Hyères Cedex. francois.lecuire@chu-lyon.fr
Benareau
I
I
Rubini
J
J
Basso
M
M
D000328:Adult; D006617:Hip Dislocation / Q000209:etiology*; D006622:Hip Prosthesis; D006801:Humans; D008875:Middle Aged; D011474:Prosthesis Design; D011475:Prosthesis Failure