An active electrode for biopotential recording from small localized bio-sources.
BACKGROUND
Laser bio-stimulation is a well-established procedure in Medical Acupuncture. Nevertheless there is still a confusion as to whether it works or the effect is just placebo. Although a plethora of scientific papers published, showing positive clinical results, there is still a lack of objective scientific proofs about the bio-stimulation effect of lasers used in Acupuncture. The objective of this work was to design and build a body surface electrode and an amplifier for biopotential recording from acupuncture points, considered here as small localized bio-sources (SLB). The design is aimed for studying SLB potentials provoked by laser stimulus, in search for objective proofs of the bio-stimulation effect of lasers used in Medical Acupuncture.


METHODS
The active electrode presented features a new adjustable anchoring system and fractionation of the biopotential amplifier between the electrode and the cabinet's location. The new adjustable electrode anchoring system is designed to reduce the electrode-skin contact impedance, its variation and motion artifacts. That is achieved by increasing the electrode-skin tension and decreasing its relative movement. Additionally the sensing element provides local constant skin stretching thus eliminating the contribution of the skin potential artifact. The electrode is attached to the skin by a double-sided adhesive pad, where the sensing element is a stainless steel, 4 mm in diameter. The fractionation of the biopotential amplifier is done by incorporating the amplifier's front-end op-amps at the electrodes, thus avoiding the use of extra buffers. The biopotential amplifier features two selectable modes of operation: semi-AC-mode with a -3 dB bandwidth of 0.32-1000 Hz and AC-mode with a bandwidth of 0.16-1000 Hz.


RESULTS
The average measured DC electrode-skin contact impedance of the proposed electrode was 450 kOmega, with electrode tension of 0.3 kg/cm2 on an unprepared skin of the inner forearm. The peak-to-peak noise voltage measured at the amplifier output, with input terminals connected to common, was 10 mVp-p, or 2 microVp-p referred to the input. The common-mode rejection ratio of the amplifier was 96 dB at 50 Hz, measured with imbalanced electrodes' impedances. The prototype was also tested practically and sample records were obtained after a low intensity SLB laser stimulation. All measurements showed almost a complete absence of 50 Hz interference, although no electrolyte gel or skin preparation was applied.


CONCLUSIONS
The results showed that the new active electrode presented significantly reduced the electrode-skin impedance, its variation and motion artifact influences. This allowed SLB signals with relatively high quality to be recorded without skin preparation. The design offers low noise and major reduction in parts, size and power consumption. The active electrode specifications were found to be better or at least comparable to those of other existing designs.
Department of Medical Physics, University of Patras, Patras 26500, Greece. emil@bit.med.upatras.gr
Emil S
ES
Valchinov
Nicolas E
NE
Pallikarakis
D015669:Acupuncture Points; D004566:Electrodes; D005071:Evoked Potentials; D006801:Humans; D007834:Lasers; D012879:Skin Physiological Phenomena