Use of EPID for leaf position accuracy QA of dynamic multi-leaf collimator (DMLC) treatment.
We describe in this paper an alternative method for routine dynamic multi-leaf collimator (DMLC) quality assurance (QA) using an electronic portal imaging device (EPID). Currently, this QA is done at our institution by filming an intensity-modulated radiotherapy (IMRT) test field producing a pattern of five 1-mm bands 2 cm apart and performing a visual spot-check for leaf alignment, motion lags, sticking and any other mechanical problems. In this study, we used an amorphous silicon aS500 EPID and films contemporaneously for the DMLC QA to test the practicality and efficacy of EPID vis-à-vis film. The EPID image was transformed to an integrated dose map by first converting the reading to dose using a calibration curve, and then multiplying by the number of averaged frames. The EPID dose map was then back-projected to the central axis plane and was compared to the film measurements which were scanned and converted to dose using a film dosimetry system. We determined the full-width half-maximum (FWHM) of each band for both images, and evaluated the dose to the valley between two peaks. We also simulated mechanical problems by increasing the band gap to 1.5 mm for some leaf pairs. Our results show that EPID is as good as the film in resolving the band pattern of the IMRT test field. Although the resolution of the EPID is lower than that of the film (0.78 mm/pixel vs 0.36 mm/pixel for the film), it is high enough to faithfully reproduce the band pattern without significant distortion. The FWHM of the EPID is 2.84 mm, slightly higher than the 2.01 mm for the film. The lowest dose to the valley is significantly lower for the EPID (15.5% of the peak value) than for the film (28.6%), indicating that EPID is less energy independent. The simulated leaf problem can be spotted by visual inspection of both images; however, it is more difficult for the film without being scanned and contrast-enhanced. EPID images have the advantage of being already digital and their analysis can easily be automated to flag leaf pairs outside tolerance limits of set parameters such as FWHM, peak dose values, peak location, and distance between peaks. This automation is a new feature that will help preempt MLC motion interlocks and decrease machine downtime during actual IMRT treatment. We conclude that since EPID images can be acquired, analyzed and stored much more conveniently than film, EPID is a good alternative to film for routine DMLC QA.
Memorial Sloan-Kettering Cancer Center, New York, New York 10021, USA. changj@mskcc.org
J
J
Chang
C H
CH
Obcemea
J
J
Sillanpaa
J
J
Mechalakos
C
C
Burman
D019544:Equipment Failure Analysis / Q000295:instrumentation / Q000379:methods; D005373:Film Dosimetry / Q000379:methods; D011785:Quality Assurance, Health Care / Q000379:methods*; D011856:Radiographic Image Enhancement / Q000379:methods*; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000295:instrumentation / Q000379:methods*; D011874:Radiometry / Q000295:instrumentation* / Q000379:methods*; D011879:Radiotherapy Dosage; D020266:Radiotherapy, Conformal / Q000295:instrumentation* / Q000379:methods*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity