title: [Transurethral resection of bladder tumours in sodium chloride solution].
abstract: BACKGROUND
  Electrical transurethral resection is a well established and developed procedure. It is performed using high frequency current in a nonconductive irrigation medium. Due to these features this procedure is compromised by two possible complications: (1) spontaneous contractions of adductors caused by electrical stimulation of the obturator nerve may lead to bladder perforation, and (2) excessive flushing of the irrigation medium into the circulation can cause TUR syndrome. We present our initial experiences with a new system for transurethral resection which has overcome these potential sources of complications. The system, developed by Olympus, works with a modified guided high frequency current in 0.9% saline as irrigant. It is called TURIS (transurethral resection in saline).
  
  
  METHODS
  A total of 35 resections of bladder tumors were performed using the TURIS technique. The operations were carried out under intravenous anaesthesia without relaxation or nerve block. During resection high frequency current passes through and active electrode (resection loop) to the sheath electrode. About 90% of the current flows through the saline to the sheath of the resectoscope, only a small amount circulates through the body of the patient. The experiences of the surgeons were documented. The resected specimens were histologically examined for artificial thermal changes and compared with a control group of conventionally resected patients.
  
  
  RESULTS
  The handling of the TURIS resectoscope is very similar to that of well known instruments. Therefore, no special training was required for the surgeon or the theatre nurse. The control of the cutting was very efficient. Coagulation of bleeding was very good. The lack of carbonisation at the resection ground led to an excellent assessment of the resected areas. The resected tissue did not stick to the resection loop. Contraction due to nerve stimulation was not observed. No complications occurred. Histology showed no significant differences in the quantity or quality of thermal artifacts due to current.
  
  
  CONCLUSIONS
  TURIS can be performed safety without a learning curve. Especially in TUR-B, it seems to be advantageous due to the excellent control of the extent of cutting and the lack of nerve stimulation. Quantity and quality of thermal changes in histology are not different from a conventionally resected control group.
authors.affiliation: Abteilung für Urologie, Allgemeines Krankenhaus Barmbek, Hamburg. c.brunken@freenet.de
authors.forename: C
authors.initials: C
authors.lastname: Brunken
authors.forename: H
authors.initials: H
authors.lastname: Qiu
authors.forename: R
authors.initials: R
authors.lastname: Tauber
mesh_terms: D000368:Aged; D000369:Aged, 80 and over; D017115:Catheter Ablation / Q000009:adverse effects / Q000295:instrumentation* / Q000379:methods*; D019544:Equipment Failure Analysis; D005260:Female; D006801:Humans; D008297:Male; D008875:Middle Aged; D010865:Pilot Projects; D012965:Sodium Chloride / Q000008:administration & dosage*; D020728:Transurethral Resection of Prostate / Q000009:adverse effects / Q000295:instrumentation* / Q000379:methods*; D016896:Treatment Outcome; D001749:Urinary Bladder Neoplasms / Q000628:therapy*
chemical_list: D012965:Sodium Chloride
vernacular_title: Transurethrale Resektion von Blasentumoren in Kochsalzlösung.